China World Technology Medical Equipment Service Group
|Minimum Order Quantity:
|3-5 working days
|T/T, Western Union，PayPal
|Nihon Kohden Defibrillator
|Medical Equipment,ICU,hospital Device
|Mode Of Transport:
|50/60 Hz ~
|Professional Technical Team
|Hospital, Clinic, Hospital Room, ICU, Sickroom, Ward, Bed-ward, Inpatient Ward
|50, 25, 5mm/s
TEC-7721C Defibrillation Printer Recorder Unit,
Defibrillation Printer Recorder Unit WS-761V,
TEC-7621C Defibrillation Printer Recorder Unit
NIHON KOHDEN Cardiolife TEC-7621C TEC-7721C defibrillation printer Recorder Unit WS-761V
► Brief Introduction
This printer Recorder Unit WS-761V is 3-channel built in thermal array recorder records various defibrillation information. And work for NIHON KOHDEN Cardiolife TEC-7621C TEC-7721C defibrillation.
► Remove the recorder unit
► Record Type
► Check Recorder Screen
You can check the recorder. When “4.Check Recorder” on the System Maintenance - Check Hardware screen in selected and the OK Button is pressed, the recorder prints test patterns.
After all test patterns are printed, the System Maintenance - Check Hardware screen is displayed.
|Printing is blurred. Dots are missing
|The specified recording papar is not used
|Use the specified recording paper.
|The thermal head is dirty.
|Clean the thermal head with the head cleaning pen. If the troble not solved. The recorder unit is faulty. Replace the recorder unit.
|There is no printing
|The recording paper is not loaded.
|Load new recording paper.
|Recording unit door is not properly closed.
|Close the door until it clicks.
|The recording paper is set with the wrong side facing up.
|Set the recording paper correctly.
|Faulty key board.
|If the lamp dose not light when you press the record key, cleck the key function and recorder with the System Maintenance screen. If the recorder check is passed, the key board is faulty. Replace the key board.
|If the lamp lights when you press the record key, check the recorder in the System Maintenance screen.
|Faulty main board
|Printing is light.
|The recording unit temperature is too hot.
|Cool the instrument down in cooler place. If the trouble is not solved, the recorder unit or main board is faulty. Replace the recorder unit or main board.
► Delivery Issues
As a general rule, all the orders are going to be shipped within 3 to 5 working days and delivered on time. Of course sometimes there are some special circumstances. For instance, on busy days the shipment will be postponed. But no worries, all the orders will be dispatched finally. We will make great efforts to process the orders and ship them out as soon as possible. Please stay patient and keep in touch with us. If you need the product to be shipped urgently, please make sure you talk to us in advance before placing your orders. In order to make the deal runs successfully, please make it a point and manage your time of placing orders on the basis of your business needs. If you have any questions such as your order is taking much more time than expected or you need to know when will your order arrive, you can contact us at anytime. We will reply you as soon as possible. Your satisfaction is our biggest wish.
If you are interested in Hospital Medical Equipment Used NIHON KOHDEN Cardiolife TEC-7621C TEC-7721C defibrillation printer Recorder Unit WS-761V , welcome to place your orders.
Contact Person: Rita