China World Technology Medical Equipment Service Group
|Minimum Order Quantity:||1pcs|
|Delivery Time:||3-5 working days|
|Payment Terms:||T/T(wire transfer), Western Union, Paypal|
|Product Name:||Philip TC20 TC30 ECG Machine Power Supply Board MODEL LPS54-M AC INPUT 100-240V-2A 50HZ 60HZ DC OUTPUT 15V=4A 60W||Brand:||Philip|
|Model:||LPS54-M||Input Voltage:||Input Voltage|
|Input Voltage Frequency:||50 To 60 Hz||Input Current:||1.2A|
|Output:||15 Volts DC, 4.0A, 60W||MOQ:||One Piece|
TC30 ECG Machine Power Supply Board,
40V ECG Machine Power Supply Board,
MODEL LPS54-M Power Supply Board
Power Supply Board MODEL LPS54-M Philipilip TC20 TC30 ECG MachineAC INPUT 100-240V-2A 50HZ 60HZ DC OUTPUT 15V=4A 60W
PageWriter TC20/TC30 CardiograPhilipilip AC Input Voltage
Input voltage: 100 to 240 VAC n
Input voltage frequency: 50 to 60 Hz
Input current: 1.2A
PageWriter TC20/TC30 CardiograPhilipilip AC Output Voltage n
Output: 15 Volts DC, 4.0A, 60W
|Item name||Philipilip TC20 TC30 ECG machine power supply board MODEL LPS54-M AC INPUT 100-240V-2A 50HZ 60HZ DC OUTPUT 15V=4A 60W|
|Use place||Hospital Devices; Medical Care; ICU;|
|Deliver time||3-5 days|
When the AC power cord is plugged into an active electrical outlet, make sure that it is also properly connected to the electrocardiograPhilipilip. When the AC power cord is not connected to the electrocardiograPhilipilip, be sure to unplug the AC power cord (mains lead) from the electrical outlet.
Use only grounded electrical wires (three-wire power cords with grounded connectors) and grounded electrical wires labeled "hospital only" or "hospital grade." Never connect a grounded plug to a grounded electrical outlet by removing the mound. Use equipotential terminals when redundant landing in accordance with IEC 60601-1-1 is required.
The power supply may be hot to the touch.
n To disconnect the electrocardiograPhilipilip from alternating current, unplug the electron current wire from the electrocardiograPhilipilip of the AC power source.
Udstyr This equipment complies with the leakage current limits specified in UL 60601-1: 2003 medical electrical equipment - general safety requirements only when connected to a 120 volt electrical source.
n Periodically inspect patient data cables, power cords, and AC power cords for damaged or cracked insulation. Make sure there are no visible wires on the AC power cord.
n Use only the Philipilip Medical Systems AC power cord supplied with the electrocardiograPhilipilip. The use of any other power source has not been tested and may result in injury to the operator or patient, including electric shock. Regularly inspect the AC power cord and AC outlet
to ensure that both are safe and functional. If the AC power cord or AC plug is not in a safe or functional condition, use the electrocardiograPhilipilip for battery power and contact Philipilip Medical Systems for repair.
Warranty: 90 days
New and used units are sold with a warranty, the length of which is determined by the type of equipment and agreed at the point of sale.
1.What's your MOQ
Our MOQ is one piece
2.May i have a sample first ?
3.What's your payment method ?
T/T ,Paypal ,Western union
4.May i have your price list ?
I'm sorry we don't have price list as our stock is changing everyday .Pls check our website and send back picture to get price .
Contact Person: Andy