China World Technology Medical Equipment Service Group
|Model Number:||SH330 SH360|
|Minimum Order Quantity:||1|
|Delivery Time:||3-5 working days|
|Payment Terms:||T/T, Western Union，PayPal|
|Features:||Professional, Reliable,Economic||Product Name:||Humidifier|
|Type:||SH330 SH360||Application:||Medical Equipment,ICU,hospital Device|
|Warranty:||90days||Temperature Display Range:||10 – 70 °C|
|Gas Temperature Measurement Accuracy::||± 2 °C||Audio Information Signal::||Single Beep Or Double Beep|
|Rated Power::||220 VA||Mains Supply::||115 V~, 60 Hz|
|Heater-plate Output::||150 W|
SH360 Respiratory Humidifier,
JIKE SH330 Respiratory Humidifier
JIKE SH330 SH360 Respiratory Humidifier Medical Equipment , ICU, Hospital Device
JIKE SH330 SH360 Respiratory Humidifier Medical Equipment
The respiratory humidifier is an important medical device connected between the ventilator and the patient. Its function is to warm and humidify the inhaled gas. Good warming and humidification can prevent and reduce secondary infection of the respiratory tract in mechanically ventilated patients, as well as stimulation of the cardiopulmonary system and keep the alveoli moist. At the same time, it can reduce the consumption of heat and moisture in the respiratory tract, so that the airway is not easy to produce phlegm scabs, and can reduce the viscosity of secretions and promote expectoration.
The principle of the medical airway humidifier is to heat and humidify the air flow by heating and humidifying the air flow, so that the air flow passes through the hot water surface to be heated and humidified, and the heating temperature is controlled by setting the temperature control device, and this temperature can be monitored through the digital display. Using a humidifier can reduce the stimulation of mechanical ventilation to the cardiopulmonary system and improve the adaptability of patients to mechanical ventilation; keep the alveoli moist, enhance their activity, and promote gas exchange; keep the airway and airway secretions moist, which is conducive to sputum suction and Prevent airway blockage.
|Inlet Port/Outlet Port；||22mm OD / 15mm ID|
|Filling Hole:||4.3mm ID|
|Material Quality:||Ploysulfone, Aluminum, Silica Gel|
|Max Volume (Max)||350mL|
|Pressure Drop||≤0.2kPa(the flow is 60L/min)|
The temperature of the ventilator humidifier is adjusted to 37-41 ℃.
1. The inspiratory filter must be installed on the ventilator end, and then connected to the humidifier.
2. The humidifier usually has a separate power cord and power switch. Be sure to turn it on after plugging in the power supply, otherwise the gas entering the patient's trachea will be dry and cold. On the contrary, it is necessary to observe and replenish the water level of the humidification tank in time to avoid overheating of the gas entering the patient's trachea caused by dry burning.
3. There may be a small temperature sensor socket on the humidifier. Be sure to plug in a special temperature sensor, otherwise it will be prone to air leakage and temperature monitoring errors.
4. If an artificial nose is used without a heated humidifier, the artificial nose must be installed between the "Y" tube and the patient's tracheal tube or tracheostomy tube.
If you want to explore more details about JIKE SH330 SH360 Respiratory Humidifier Medical Equipment , please let us know. Looking forward to receiving your early inquiry.
Contact Person: Rita