China World Technology Medical Equipment Service Group
|Model Number:||P/N M4046-60001|
|Minimum Order Quantity:||1pcs|
|Delivery Time:||3-5 working days|
|Payment Terms:||T/T(wire transfer), Western Union, Paypal|
|Product Name:||P/N M4046-60001 Philip MP70 Patient Monitor AC/DC Power Supply Module||Brand:||PhilipLIPS|
|Part No .:||M4046-60001||Condition:||Refurbished|
|Power Consumption:||<145W||Line Voltage:||100-240V~|
|Warranty:||90 Days||Delivery Method:||DHL,UPS ,Fedex|
Philip MP70 Patient Monitor Parts,
M4046-60001 Patient Monitor Parts,
M4046-60001 DC Power Supply Module
P/N M4046-60001 Philipilip MP70 Patient Monitor AC/DC Power Supply Module
Philipilip MP70 patient monitor power supply module M4046-60001
The AC/DC converter converts the AC power from the socket to 48V/120W DC and isolates the monitoring system from the AC power source. 48V is distributed through the busbar and supplies all system components: The 56V DC power required to extend the FMS, MMS and metering servers is provided by an isolated DC/DC converter. The power required for the backlight is converted to 12V DC by the DC/DC backlight converter. This processor is powered by a voltage of 3.3 V and 5 V DC. The conversion takes place in two steps: The first DC/DC converter is a power conditioner which reduces fluctuations caused by changes in load on the 48V power strip. Another DC/DC converter converts the current into the required voltage. The interface card requires 10V AC power. The HIF and LED cards are powered by an unregulated 12V DC power supply. The integrated module slot requires 5V for the module slot and 60V when 48V and another DC/DC converter is used to supply the module with
Removing Power Supply
When replacing the power supply, the monitor must be charged with the serial number. Auxiliary tools are required for this task. For details on how to submit a new serial number, see Instructions for Using Support Tools.
1. Remove the cable management cover and white plastic cover as described in Removing the I / O board.
2. Remove the top cover on the back of the monitor. It is divided into two parts, the middle part is divided first.
3. Rotate the monitor towards you.
4 .Disconnect the colored power cord.
5. Use a T20 screwdriver to remove the two screws (one at the top and one at the bottom)
6. Move the power supply to the top of the monitor and remove it.
Installation NOTE: When reinstalling, please switch the power to the bottom of the monitor (from step 6 above).
1. How long is the delivery?
Delivery time is around 2-3 working days ,and around 5-7 working days to arrive on your side
2. What’s your warranty ?
Our warranty is 90 days ,for some new products ,we can offer one year warranty .For any problems during our warranty ,we can replace another one .
3. What’s your payment terms ?
Our payment terms is full before the shipment .
4. How to place the order ?
Contact us and confirm order details ,once confirmed ,proforma invoice will be send back and shipment will be arranged after receiving full payment .
If you have any interest in P/N M4046-60001 Philipilip MP70 Patient Monitor AC/DC Power Supply Module ,feel free to contact us .
Contact Person: Andy