China World Technology Medical Equipment Service Group
|Model Number:||M8048A FMS-8|
|Minimum Order Quantity:||1pcs|
|Delivery Time:||3-5 working days|
|Payment Terms:||T/T(wire transfer), Western Union, Paypal|
|Product Name:||M8048A FMS-8 Philip MP60 MP70 Patient Bedside Monitor Module Rack||Brand:||Philip|
|Part No.:||M8048A FMS-8||Weight:||<3500g ,<7.7 Lb|
|Weight*Height *Width:||<315mm*130mm*120mm (12.4*5.1*4.7inch )||Warranty:||90 Days|
|Condition:||New & Original||MOQ:||One Piece|
|Payment Terms:||Full Before Shipment||Payment Method:||T/T ,Paypal ,Western Union|
MP70 Patient Monitor Module Rack,
M8048A Patient Monitor Module Rack
M8048A FMS-8 Philipilip MP60 MP70 Patient Bedside Monitor Module Rack
Philipilip FMS module rack
MAX weight:<3500g ,<7.7 lb
Weight*Height *Width:<315mm*130mm*120mm (12.4*5.1*4.7inch )
Flexible Module Server Troubleshooting
|Symptoms||Cause of Failure||Failure Isolation and Remedy|
|Prompt Message “Unrecognized Measurement Module in slot s” is issued||An unsupported module has been plugged into the Flexible Module Server||Unplug the unsupported module|
|Prompt message “Measurement Module in slot n is currently ignored” is issued||Too many modules of the same kind have been plugged into the Flexible Module Server||Unplug module in slot n|
|Red Error LED stays on||Unrecoverable hardware selftest error:||Try connecting the MMS directly to the MSL cable. If the measured value is displayed on the screen, the fault is in the FMS|
|MSL cable defective||If no measurements are displayed when the MMS is directly connected to the MSL cable, replace the MSL cable|
|Flex connector from main board to MSL defective||Replace connector|
|CPU module defective||Replace CPU module|
|Red Error LED flashes||Hardware selftest error||If system comes up, check status log. Otherwise see above|
|Flexible Module Server LEDs OK, Front Measurement Module not recognized (silent or INOP)||Damaged measurement module or measurement module connector||Replace Measurement Module|
|No Front-End power because MSL voltage from the monitor is too high or too low||
1. Try a new MSL cable. Replace if failure is rectified.
2. Replace mother board
|Mother board or connector on Mother Board defective||If the voltage is within range, or if the connector is visibly damaged, replace the motherboard|
Philipilip FMS/MMS Module Lock
The following parts and hardware are included in this mounting kit
|3||#8-32 x 1/4'' Button Head Machine Screw, Tamper-proof||4|
|4||5/32'' Hex Wrench, Tamper-proof||1|
Mounting Notes: The bracket is designed to accommodate FMS (handle on both ends) or MMS extension (handle on one end). Installation is only shown on FMS with MMS extension.
Installing the Module Lock
1. After installing the required modules, slide the FMS into the main housing as shown below. Installation Note: It is not necessary to remove the previously installed mounting clips from the FMS to install the module lock.
2.Slide the support bracket over the MMS extension and into the main housing as far as it will go. Note the slot on the bracket for mounting the module lock on the second handle.
3.Using the 5/32'' tamper-resistant hex key provided, install four (4) #8-32 x 1/4'' tamper-resistant screws as shown below.
Installation Notes: Module locks are designed to provide space for individual modules to be disconnected without removing them from the server.
Contact Person: Andy