China World Technology Medical Equipment Service Group
|Brand Name:||COVIDIEN Nellcorr|
|Minimum Order Quantity:||1|
|Delivery Time:||3-5 working days|
|Payment Terms:||T/T, Western Union，PayPal|
|Product Name:||COVIDIEN Nellcorr Pulse Oximeter||Type:||N560|
|Mode Of Transport:||Express Delivery||Temperature Operating:||50 ºF To 113 ºF (10 ºC To 45 ºC)|
|Altitude/Barometric Pressure:||-390 M To 3,012 M (-1,280 Ft. To 9,882 Ft.) 70 KPa To 106 KPa (20.6 In. Hg To 31.3 In. Hg)||Nominal Voltage:||100 To 240 Volts AC, 25 To 39 Volt/amps To Be Compliant With IEC 60601-1 Sub-clause 10.2.2|
|Weight:||3.07 Lbs. (1.39 Kg) Without Pole Mount||Dimensions:||2.95 " (H) × 9.06 " (W) × 5.04" (D) (75 Mm (H) × 230 Mm (W) × 128 Mm (D)|
|Fuses:||Qty 2, 2 A, 250 Volts, Slow-blow, IEC (5 X 20 Mm)||Function:||Monitoring Patient's Health Condition|
|Apply For:||Hospital, Clinic, Hospital Room, ICU, Sickroom, Ward, Bed-ward, Inpatient Ward|
COVIDIEN Nellcorr Pulse Oximeter,
N560 Pulse Oximeter
Hospital Medical Equipment COVIDIEN Nellcorrr OxiMax N560 N-560 Pulse Oximeter
► Brief Introduction
The N-560 can be used for patients of all ages — infants, children, and adults. The N-560 continuously measures the patient’s pulse rate and the percentage of oxygen circulating in the blood. When either the patient’s pulse rate or the percentage of oxygen goes below or above a pre-set alarm limit, the N-560 warns you by sounding an alarm, lighting an indicator, or changing the number display from green to red.
► Front Panel Control Buttons
1. Power On/Off Button
The Power On/off button is used to turn the N-560 on or off. The Power On/Off button must be pressed and held for approximately one second in order to turn the N-560 on or off.
2. Alarm Silence Button
The Alarm Silence button is used to silence current audible alarms for the alarm silence duration period. When an alarm has been silenced, pressing the button again reactivates, or “unsilences” the alarm.
3. Adjust Up Button
The Adjust Up button is used to increase variable settings of the N-560.
4. Adjust Down Button
The Adjust Down button is used to decrease variable settings of the N-560.
5. Pulse Rate Alarm Limit Button
The Pulse Rate Alarm Limit button is used to view the pulse rate alarm limit. The Adjust Up and Adjust Down buttons are used to increase or decrease the alarm limit setting. The clinician will determine if these settings need to be changed.
6. SatSeconds Alarm Limit Button
The SatSeconds™ Alarm Limit button is used to view the SatSeconds alarm limit. SatSeconds is an alarm management system. The clinician will determine if this feature is appropriate for the patient. The Adjust Up and Adjust Down buttons are used to change the SatSeconds limit settings.
7. SpO2 Alarm Limit Button
The SpO2 Alarm Limit button is used to view the SpO2 alarm limit. The Adjust Up and Adjust Down buttons are used to increase or decrease the alarm limit setting. The clinician will determine if these settings need to be changed.
► Front Panel Status Indicators and Components
|1||SpO2 Port (DOC-10 Cable Connection)|
|2||Low Battery Indicator|
|3||AC Power Indicator|
|4||Alarm Silence Indicator|
|5||Sensor Message Indicator|
|6||Sensor Off Indicator|
|7||Data In Sensor Indicator|
|9||Pulse Search Indicator|
|10||Speaker (Audible Alarms)|
Thank you for your time and interest. All through the way we seek for providing perfect service to the satisfaction of customers. If you are interested in COVIDIEN Nellcorrr OxiMax N560 N-560 Pulse Oximeter, welcome to place your orders.
Contact Person: Rita