China World Technology Medical Equipment Service Group
|Minimum Order Quantity:||1|
|Delivery Time:||3-5 working days|
|Payment Terms:||T/T, Western Union，PayPal|
|Product Name:||Nellcor SpO 2 Extension Cable||Type:||DEC-8|
|Application:||Medical Equipment,ICU,hospital Device||Warranty:||90days|
|Color:||Blue||Stock Situation:||Available Now|
|Service:||Online Technical Support,Passion In Serving Our Customer With The Best Service||MEANS OF TRANSPORT:||Express Delivery|
|Preponderance:||Advanced Equipment,Original||Minimum Purchase Commitment:||1pcs|
|Scenario:||ICU, Emergency, ICU Room And Operating Room,etc||Product Specialty:||Easy To Use,Economic,Functional|
|Power Supply Source:||Electricity||Power Consumption:||110V—240v|
Pulse Oximetry SpO2 Extension Cable,
SpO2 Extension Cable
Nellcor DEC-8 pulse oximetry SpO 2 extension cable for Welch Allyn Vital Signs Monitor 300 Series
► Brief Introduction
Verify that the sensor cable connector has slots on both sides. If the cable connector has a slot on only one side, then the sensor is not compatible with the monitor.
The SpO 2 connectors are notched and flanged to ensure proper fit. If the connectors do not join easily, stop and verify the following:
► Test connections
|Test connections||Connector/cable||Connect to||Wiring||Connect to|
|SpO 2 (Nellcor)||NellcorD-connector(DEC-8 or other OXI-MAX equivalent)||SpO 2 Input Connector||All wires connected together and terminated appropriately for your hi-pot tester||High Voltage|
► Testing the monitor only (Nellcor SpO 2 )
Use this procedure to test only the monitor SpO 2 function.
1. Power the monitor off.
2. Connect the Nellcor SRC-MAX SpO 2 functional tester to the SpO 2 input connector through a Nellcor DEC-8 extension cable.
3. Power the monitor on.
4. Verify the following on the SRC-MAX:
5. Give the monitor up to 30 seconds to stabilize, and verify a displayed pulse rate of 60 ± 3 bpm and a displayed SpO 2 of 75 ± 3%.
6. Set the SRC-MAX pulse rate to 200 bpm.
7. Give the monitor up to 30 seconds to stabilize, and verify a displayed pulse rate of 200 ± 3 bpm.
8. Switch the SRC-MAX SpO 2 saturation percentage to 90.
9. Give the monitor up to 30 seconds to stabilize, and verify a displayed SpO 2 saturation level of 90 ± 3%.
10. Disconnect the SRC-MAX.
► Test the isolation between SpO2 and DC input:
► Company Profile
We mainly sell different kinds of medical equipments, accessories and repair parts including patient monitor, defibrillator machine, patient monitor printer, ECG machine, EKG machine, main board, batteries and so on. After many years of marketing and developing, we are quiet proficient in medical machines technology.
Our products are increasingly popular and enjoy good reputation at home and abroad. Relying on superior products quality and reasonable price, our sales have been growing steadily with each passing year. We are a reliable manufacturer providing the best products and services for global customers. Satisfying all the customers' demands is our permanent pursuit.
Looking forward to receiving your early inquiry about Nellcor DEC-8 pulse oximetry SpO 2 extension cable for Welch Allyn Vital Signs Monitor 300 Series. If you have other questions about our products or services, please feel free to contact us. We are always ready to serve you.
Contact Person: Rita