China World Technology Medical Equipment Service Group
|Minimum Order Quantity:
|3-5 working days
|T/T, Western Union，PayPal
|Power Supply Source:
|EDAN F2 F3 F6 F9 Fetal Monitor
|Mode Of Transport:
|Minimum Purchasing Quantity:
TP01-RQ-21 TOCO Transducer,
Patient Monitor Parts TOCO Transducer,
F9 Fetal Monitor TOCO Transducer
EDAN F2 F3 F6 F9 TOCO Transducer Blue label MS3-31527(B) TPU REF 02.01.210259 MPN02.01.210259012 (6)06944413805507 (21)H18900830290
► Brief Introduction
1) Placing Transducer Belt
Place the transducer belts across the bed, ensuring that the belt will be around the abdomen when it is fastened. Lay the patient on the bed.
Alternatively, the patient can take a sitting position. Arrange the belt around her abdomen.
2) Fixing the Transducer
Wipe any gel remaining on abdomen around this area. Place the TOCO transducer on the patient’s abdomen, which is flat and approximately 3 cm away from the fundus, e.g. slightly above the umbilicus on the left or on the right. The position should be different for different purposes: place the transducer close to the fetal buttocks for NST, and place it on fetal back in delivery.
Wrap the abdomen with the belt over the transducer. Fix the transducer by pushing its buckle through the overlapping section of the belt. Make sure the belt fits the patient snugly and comfortably.
3) Adjusting the Numeric to Zero
Press the AUTO ZERO key to adjust the numeric to the baseline. Make sure this is not done during a contraction.
Wipe off any gel presents on abdomen around this area
|Baseline Drift due to Temperature Changes:
|≤ 1 unit/min/°C (free air)
|≤ 5 units/min/°C (underwater)
► Relocation of the Transducers
Transducers may be belted on the patient for a long time without stop. In rare cases, this may lead to irritations to the patient skin. To avoid skin irritations, please inspect the application site at least every three hours. If the skin quality changes, you should move the transducer to another site.Please periodically inspect the application site (between contractions) of TOCO transducer at least every three hours.
To reduce the risk of skin irritations, do not allow residual cleaning agent or disinfecting agent on the surface of transducers. Before using cleaning agent and/or disinfecting agent, refer to the cleaning and disinfecting sections in this user manual. Wipe the transducer surface with a cloth dampened with water before applying to the patient.
► Quality Guarantee
"Quality assurance" is our eternal pursuit and promise. We have a professional division of labor, a highly cooperative team, and a strong technical force in medical device development. Our expert team provides fast and reliable solutions for the sales and maintenance of patient monitors, defibrillators, ECG/ECG machines and other accessories or repair parts. We have always focused on producing quality products and providing first-class customer service. It is important to note that there are brand new and second-hand products for sale. Please distinguish them carefully when placing an order. Otherwise it will cause unnecessary trouble. If you have any questions about product quality, please contact us. We will get back to you as soon as possible.
Thank you for your time! If you have other questions about our Edan F2 F3 F6 F9 Fetal Monitor Toco Transducer Probe IPX1 TP01-RQ-21 Compatible , please feel free to contact us. We are always ready to serve you.
Contact Person: Rita