China World Technology Medical Equipment Service Group
|Brand Name:||Mindray D3 Rechargeable Li-ion battery|
|Minimum Order Quantity:||1|
|Delivery Time:||3-5 working days|
|Payment Terms:||T/T, Western Union，PayPal|
|Product Name:||Mindray Rechargeable Lithium-ion Battery||Type:||0146-00-0099|
|Rated Battery Capacity::||4600mAh||Battery Run Time: (one Battery):||>130 Min|
|Battery Run Time:(two Batteries):||>270 Min||Battery Charge Time:powered Off:||3.5 Hours For 90% Charge And 5.5 Hours For Full Charge|
|Battery Charge Time:powered On:||5.5 Hours For 90% Charge And 8.5 Hours For Full Charge|
Patient Monitor Lithium Ion Battery,
Rechargeable 11.1V Lithium Ion Battery,
0146-00-0099 Lithium Ion Battery
Mindray Datascope Passport V Patient Monitor Rechargeable Lithium-ion Battery 11.1V 4600mAh REF 0146-00-0099
► Brief Introduction
This battery power is introduced to the power module via the battery interface board, and then processed and distributed to each component by the power management and interface board.The battery compartment door is located at the left-bottom of the patient monitor.Storage of the lithium-ion batteries depends on temperature, time period and the degree of cell charging state. After 6 months of storage at 23oC, fully charged lithium-ion batteries have a retention capacity of 93%.
► Battery Replacement
1. Open battery compartment door, on left side of unit, by pressing the finger grip area and sliding the door to the rear.
2. Slide the release located at the top of the installed battery. This will eject the battery. Slide out and remove battery.
3. Slide in replacement battery until it clicks into place.
4. Close battery compartment door by sliding the door to the right until it firmly clicks into place.
► Battery Status:
When batteries are installed and the monitor is functioning on battery power, the battery indicator provides a visual reference for the approximate charge level of the batteries. See the following examples.
A green LED located above the battery icon indicates that the battery charger is active. The charger will not always be active when AC power is present. It is dependent on the battery charge condition. The LED is not an indication of the condition of the batteries or their charge level. Charged batteries must be installed in the monitor to ensure uninterrupted operation while switching from AC to battery power.
When the battery charge is low, but not below the cutoff voltage, a flashing low battery icon is displayed and a special low battery sound is activated.
|Rated Battery capacity:||4600mAh (one battery)|
|Battery run time:||>270min (two batteries),
>130 min (one battery)
|Battery charge time:||3.5 hours for 90% charge and 5.5 hours for full charge (equipment powered off)
5.5 hours for 90% charge and 8.5 hours for full charge (equipment powered on)
► Product Use
We specialize in the sales and maintenance of various medical equipment and accessories. Our main brands are Philips, Mindray, GE, Goldway, NIHON KOHDEN, etc. Since the establishment of the company, we have explored and developed in this industry for many years. Our products are mainly used for medical equipment in hospitals or clinics. We sell complete machines, spare parts, and also develop product maintenance services. Our main products are patient monitors, defibrillators, ECG machines and various module accessories. If you want to know more detailed information about our products, welcome to contact us. We will give you a detailed introduction.
Customer satisfaction. Our biggest pursuit! If you have any questions about Mindray Datascope Passport V Patient Monitor Rechargeable Lithium-ion Battery 11.1V 4600mAh REF 0146-00-0099, please feel free to contact us at any time, we are glad to answer them.
Contact Person: Rita