China World Technology Medical Equipment Service Group
|Minimum Order Quantity:||1|
|Delivery Time:||3-5 working days|
|Payment Terms:||T/T, Western Union，PayPal|
|Application:||Medical Equipment,ICU,hospital Device||Warranty:||90days|
|Color:||White||Management Principle:||Sincere And Trustworthy|
|Inventory Status:||Stock Available||MEANS OF TRANSPORT:||Express Delivery|
|Minimum Purchase Commitments:||1unit||Power Supply Source:||Electricity|
|Nominal Voltage:||110V—240v||Product Condition:||In Good Shape|
Mindray IMEC8 Patient Monitor Parts,
Patient Monitor Parts Connector Panel Board,
Patient Monitor Parameter Connector Panel Board
Hospital Medical Equipment Parts Mindray iMEC8 Patient Monitor parameter Parameter Connector Panel Assembly panel board
► Brief Introduction
The multi-parameter module, as well as the NIBP pump and electromagnetic valve, are located on the bracket of the multi-parameter module, which is fixed to the rear housing with 4 screws.
The multi-parameter module is connected to the connector board with a 2 x 7 pin cable to supply power for the parameter module, communicate with the main board, and realize analog output and Defib sync output. It adopts the DSP processor as the core. For ECG applications, the self-developed Mindray ECG_ASIC chip is used to realize 3/5/12-lead ECG, and RESP (I/II lead selectable) measurement, which greatly simplifies the sampling and control of ECG signals for the DSP.
As for NIBP, IBP, and TEMP measurement, the high-precision Σ - Δ ADC is used to realize a direct sampling of corresponding parameter signals. Mindray SpO2 measurement is realized by a stacking connector. The Nellcor and Masi-mo SpO2 boards are also compatible with the multi-parameter module, being connected through a BTB socket.
► Removing the Parameter Connector Panel Assembly
1. For equipments with standard configuration, the parameter connector assembly can be removed directly. Disconnect the multi-parameter signal cable, NIBP pipe, and ECG signal cable as shown below. Then remove the parameter connector assembly.
2. For equipments with optional C.O. and CO2 modules, remove the parameter module assembly.
► Removing the Cables From the Parameter Connector Panel
To remove the cables, loose the connectors clockwise with a sharp nose plier; To reassemble the cables, screw the connectors counter-clockwise to the correct sockets as shown below
► Environmental Specification
|Item||Operating conditions||Storage conditions|
|Temperature (℃)||0 to 40 ℃ ( 32 to 104℉ )||-20 to 60|
|Relative Humidity||15% to 95%||10% to 95%|
|Barometric (mmHg)||427.5 to 805.5||120 to 805.5|
► Warranty Issues
Under normal circumstances, both our new and old products usually have a three-month warranty. In other words, if you find any problems with the product you buy, you can replace or repair the machine during the warranty period. But please take it easy. We hope you are satisfied with our products and services. In addition, we have great confidence to make you satisfied, because our goods are of high quality. Years of sales experience enable us to deal with this kind of problem skillfully. Customer satisfaction is our goal.
f you have any questions about Mindray iMEC8 Patient Monitor parameter Parameter Connector Panel Assembly panel board, please feel free to contact us at any time, we are glad to answer them. Thank you for your time!
Contact Person: Rita