China World Technology Medical Equipment Service Group
|Minimum Order Quantity:||1|
|Delivery Time:||3-5 working days|
|Payment Terms:||T/T, Western Union，PayPal|
|Application:||Medical Equipment,ICU,hospital Device||Warranty:||90days|
|Power Consumption:||110V—240v||Product Conditions:||In Good Order|
|Power Supply Source:||Electricity||Management Principle:||Original Assembled Real Product, Quality Guarantee, Deliver On Time, Preference Price|
|Inventory Status:||Sufficient Inventory||Peculiarity:||Professional,Useful，Durable In Use|
|Product Applications Scenarios:||Hospital, Clinic, Hospital Room, ICU,etc||MEANS OF TRANSPORT:||Expressage|
|Minimum Purchase Commitments:||1 Pcs||Tag:||Battery Compartment Assembly|
Mindray T5 Patient Monitor Battery,
6802-30-66763 Patient Monitor Battery
Mindray T5 Patient Monitor Battery Compartment Assembly 6802-30-66763
The patient monitor uses two chargeable lithium-ion batteries (11.1 V, 4500 mAh). The battery compartment is located at the bottom of the patient monitor. The battery power is supplied to the mother board via the battery interface board, and then to the power module
► Exploded View
|2||Battery interface board|
|3||Crosshead screw M3×6|
|4||Conducting block 9.5×3.2|
|6||Battery latch fixture|
|7||Crosshead screw M3×10|
|9||Crosshead screw M3×6|
► Removing Battery Compartment Assembly
Unplug the cable that connects the battery compartment assembly and the mother board. Then, unscrew the three M3×6 screws and take out the battery compartment assembly.
► Conditioning a Battery
A battery needs at least two conditioning cycles when it is put into use for the first time. A battery conditioning cycle is one complete, uninterrupted charge of the battery, followed by an uninterrupted discharge of the battery. Batteries should be conditioned regularly to maintain their useful life. Condition the batteries once when they are used or stored for two months, or when their run time becomes noticeably shorter.
To condition a battery, follow this procedure:
1. Disconnect the patient monitor from the patient and stop all monitoring and measuring procedures.
2. Insert the battery in need of conditioning into the battery slots of the patient monitor.
3. Apply AC power to the patient monitor and allow the battery to charge uninterruptedly for above 6 hours.
4. Remove AC power and allow the patient monitor to run from the battery until it shuts off.
5. Apply AC power again to the patient monitor and allow the battery to charge uninterruptedly for above 6 hours.
6. This battery is now conditioned and the patient monitor can be returned to service.
► Order Information
To make sure the deal runs successfully and efficiently, please learn about and agree to the following clauses and terms. There are mainly two important things that you need to know well. The first thing is that you’d better bear in mind to look through all the product pictures with patience when purchasing so that you can confirm the item you are viewing is just what you are looking for and need to buy. It is absolutely necessary to do so as product pictures also play essential roles in product description, which helps you understand the product more intuitively and detailedly. The second thing comes to the customs fees. Please kindly know that the buyer is responsible for the customs fees such as all taxes and duties. Consequently, before paying for orders you must check out that the equipment your are ordering is allowed to imported into your country. That is to say, remember to read all the relevant information before ordering.
"Quality assurance" is our eternal pursuit and promise. Don't be hesitant to contact us! Welcome to place your orders of Mindray T5 Patient Monitor Battery Compartment Assembly 6802-30-66763. Any inquiry will be highly appreciated and thanks for your support.
Contact Person: Rita