China World Technology Medical Equipment Service Group
|Brand Name:||Fukuda Denshi|
|Model Number:||CardiMax FX-7202|
|Minimum Order Quantity:||1|
|Delivery Time:||3-5 working days|
|Payment Terms:||T/T, Western Union，PayPal|
|Product Name:||Fukuda Denshi||Type:||CardiMax FX-7202|
|Mode Of Transport:||Express Delivery||Frequency Response::||0.05 To 150Hz|
|Sensitivity::||1/4, 1/2, 1, 2cm/mV, Or Automatic||Leads::||Standard 12 Leads|
|Time Constant::||3.2 Sec.||Input Impedance::||50M Ohms Or Greater|
|Input Circuit Current::||5x10-8 A Or Lower||Sampling Rate:||8000/Sec/CH|
|Apply For:||Hospital, Clinic, Hospital Room, ICU, Sickroom, Ward, Bed-ward, Inpatient Ward|
Fukuda Denshi Patient Monitor,
CardiMax FX-7202 Patient Monitor,
ECG Machine Patient Monitor
Fukuda Denshi CardiMax FX-7202 Electrocardiograph ECG Machine Hospital Medical Equipment
► Brief Introduction
The Cardimax FX-7202 measures a standard 12-lead electrocardiogram (ECG) using advanced microprocessor technology. The FX-7202 is compact and portable. With an optional rechargeable battery, you can take it anywhere to perform a medical examination.
Displays ECG waveforms, patient information, and equipment status. Used also to enter patient data or perform various operations by touching the touch-panel keys (soft keys) provided on the screen.
Provides the controls for the Cardimax FX-7202 operation.
3.Magazine open button
Push to open the magazine cover.
Load the recording paper here.
5.Thermal print head
Records (prints) the ECG waveforms and characters, such as the measured values.
6.LAN port (10 BASE T)
Connect the LAN cable here for connection to a personal computer.
Plug in the matching end of the power supply cord here.
8.Potential equalization terminal
Connect the potential equalization cable (CE-12) to this terminal.
9.BATTERY PACK compartment
Install the optional battery pack (T8HRAAU-4713) in this compartment.
Install the optional program cartridge (PC-7403) in this compartment.
Plug in the patient cable here.
► Before attaching the electrodes to a patient
Check the condition of the patient
Check the condition of the patient and make sure they are resting quietly and comfortably. If the patient is tense, relieve tension by talking to the patient. For example, ask them to "Relax your arms and legs and feel completely at ease" or "Breath normally". You cannot record an accurate ECG if the arms and legs of the patient are tense or the patient keeps moving about. This will cause problems such as electrocardiogram interference and unstable baseline.
Cleaning the skin
Use an alcohol-moistened cloth and wipe parts of the patient's body where the electrodes are going to be attached to remove dirt and fatty oil. If foreign substances remain on the skin or an electrode, the contact resistance will increase and the recording will become unstable.
Applying Keratin cream
Apply Keratin cream for better contact between the skin and electrode. Rub the cream into the patient's skin until the skin becomes slightly reddish. When applying the cream to different parts of the chest to attach electrodes, make sure the cream on these different parts does not overlap or touch. If adjacent electrode locations on the chest become linked through Keratin cream applied too generously, an accurate ECG will not be recorded.
|Leads:||Standard 12 leads|
|Sensitivity:||1/4, 1/2, 1, 2cm/mV, or automatic|
|Polarization voltage:||±550mV or greater|
|Frequency response:||0.05 to 150Hz|
|Time constant:||3.2 sec.|
|Common mode rejection:||2mmp-p or lower (103dB or more)|
|Input impedance:||50M ohms or greater|
|Input circuit current:||5x10-8 A or lower|
|Internal noise:||20 μ Vp-p or lower|
|Calibrated voltage:||1mV±2% or lower|
Thank you for your time and interest. All through the way we seek for providing perfect service to the satisfaction of customers. If you are interested in Fukuda Denshi CardiMax FX-7202 Electrocardiograph ECG Machine Hospital Medical Equipment, welcome to place your orders.
Contact Person: Rita