China World Technology Medical Equipment Service Group
|Minimum Order Quantity:||1pcs|
|Delivery Time:||3-5 working days|
|Payment Terms:||T/T(wire transfer), Western Union, Paypal|
|Product Name:||Mindray R12 ECG Machine Repair Part Paper Tray With Good Condition||Brand:||Mindray|
|Suitable For:||Minddray R12||MOQ:||One Piece|
|Payment Method:||T/T ,Paypal ,Western Union||Payment Terms:||Full Before Shipment|
|Delivery Time:||3-5 Working Days||Delivery Method:||DHL,UPS ,Fedex ,etc|
Mindray R12 ECG Machine Parts,
Paper Tray ECG Machine Parts
Mindray R12 ECG Machine Repair Part Paper Tray With Good Condition
Load the paper
Reports can be printed from a thermal recorder or an external printer. Make sure the paper is loaded before printing the report.
The thermal insulation material supports Z-folding. Download paper:
1. Raise the level at the bottom of the tray, then pull the tray out until it stops.
2. Load a stack of paper in the tray.
3. Lift the first paper, turn it over on the roll holder, and align the top edge of the paper with the paper guides. Make sure the print side (grid side) is facing up and the black mark appears in the lower left corner of the paper.
4. Press the tray firmly until it snaps into place.
The device can print on A4 (295 mm × 210 mm) or US Letter (8.5 "× 11") paper. The paper tray is configured to match the appropriate paper size for the target location when the device is shipped from the factory.
To change the paper size, move the white plastic dividers in the tray to constrain the tray.
n For A4 size paper, insert the spacer into the slot at the top of the tray.
n For US letter size paper, insert the separator into the slot at the bottom of the tray.
|Symptom||Possible Cause||Corrective actions|
|The recorder does not work.||
5. The thermal recorder is disabled due to low battery.
1. Select Settings → [Report Settings] and disable [Paperless Log].
2. Check that the recording paper is loaded correctly.
3. Make sure the paper tray clicks into place.
4. Wait for the print head to cool down.
5. Check "Battery low! Voice recorder disabled". They are presented. If so, connect the device to AC power to operate the device and charge the battery.
|Paper jammed or misaligned.||
1. Make sure you are using the specified paper.
2. Remove the paper and tear off the jammed part. Reload the paper as described in Loading Paper.
3. Check that the paper tray spacer is set for the paper size. For details, see Loading Paper.
|Recorder door not closed||Paper tray is open.||Push the paper tray back to snap in position and try again.|
|Printer out of paper||The external printer runs out of paper||Load the paper and try again.|
Our MOQ is one piece .Welcome to take samples to test our quality .
Usually it will take around 2-3 working days to prepare for the order after payment and you will receive it around 5-7 working days .
Sure .We can ship to your Chinese forwarder free .
Sure .Video can be sent as well before the shipment .
If you're interested in Mindray R12 ECG Machine Repair Part Paper Tray With Good Condition ,feel free to contact us .Thank you .
Contact Person: Andy