China World Technology Medical Equipment Service Group
|Minimum Order Quantity:||1pcs|
|Delivery Time:||3-5 working days|
|Payment Terms:||T/T(wire transfer), Western Union, Paypal|
|Product Name:||P/N M3535-62301 Philips MRX Defibrillator Spare Parts NBP Module||Brand:||Philips|
|MOQ:||One Piece||Warranty:||90 Days|
|Delivery Time:||3-5 Working Days||Delivery Method:||DHL,UPS ,Fedex ,etc|
|Payment Method:||T/T ,Paypal ,Western Union||Packaging:||Carton|
Philips MRX Defibrillator Spare Parts,
Defibrillator Spare Parts M3535-62301,
Defibrillator NBP Module
P/N M3535-62301 Philips MRX Defibrillator Spare Parts NBP Module
The NBP module performs the following functions:
• Inflate and deflate the NIBP cuff.
• Measure the pressure in the cuff.
• Detects the pressure waveform and extracts the systolic and diastolic values from the waveform.
• Calculates mean pressure based on waveform, systolic and diastolic data
The NBP and CO2 modules are mounted on the tray. To replace any module, you must remove the tray and then replace each module.
1. Open and remove the housing.
2. Check the position of the rear housing
3. Remove the therapy capacitor.
4. Remove the NBP and CO2 module tray
Remove the NBP module and screen
A. Loosen and remove the four T-10 screws.
B. Lift the NBP module and shield out of the module tray.
1. Replace the NBP module and screen.
Place the NBP module and screen on the module tray.
2. Replace the screw and tighten.
3. Replace the NBP and CO2 module tray.
4. Replace the therapy capacitor
5. Turn off the house.
NBP Monitoring Problems
|Symptom||Possible Causes||Suggested Solution|
|NBP Equip Malfunction||NBP hardware malfunction||Replace NBP module|
|NBP Calibration Overdue||The NBP module should be calibrated once a year or every 10,000 cycles||Calibrate the NBP module|
|NBP Measurement Failed||A measurement value could not be obtained.||Check accuracy|
|NBP Service Required||The NBP module has reached its end of life, defined as 50,000 cycles.||Replace NBP module|
|Measurement cycle doesn’t start.||
|Pump operates, cuff inflates normally, but cuff will not deflate.||NBP module failure||Replace NBP module.|
Each procedure assumes that the monitor/defibrillator, pressure gauge, and inflation chamber are still configured as they were when the previous test was completed.
If all results are as indicated, the device passes this section of the test. Press the [Main Service] softkey to return to the service mode main menu.
In the event of a malfunction, start troubleshooting and repair the device as necessary.
If you're interested in P/N M3535-62301 Philips MRX Defibrillator Spare Parts NBP Module ,feel free to contact us .Thank you .
Contact Person: Andy