China World Technology Medical Equipment Service Group
|Minimum Order Quantity:||1pcs|
|Delivery Time:||3-5 working days|
|Payment Terms:||T/T(wire transfer), Western Union, Paypal|
|Product Name:||989803196521 Philip IntelliVue X3 MX100 Patient Monitor Rechargeable Lithium Ion Battery 10.8V 2000mAh||Brand:||Philip|
|Part No:||989803196521||Condition:||Original New|
|Technology:||Lithium Ion||Power:||10.8 Volt|
|Use With Philip Equipment:||Use With Philip Equipment||Single- Or Multi-patient Use:||Multi-patient|
|Sterile Or Non-sterile:||Non-sterile||Rechargeable:||Yes|
|Current:||2.0 A||Use With Non-philip Equipment:||No|
|Packaging:||1 Battery||Package Weight:||183 Gram|
|Capacity:||21.6 Wh||WxHxD:||69.6 X 72.3 X 21.6 Mm (2.7 X 2.8 X 0.8 Mm)|
MX100 Patient Monitor Accessories,
989803196521 Patient Monitor Accessories,
Patient Monitor Lithium Ion Battery
989803196521 philip IntelliVue X3 MX100 Patient monitor Rechargeable Lithium Ion Battery 10.8V 2000mAh
The battery is required for the operation of the monitor. The battery lifetime is 3 years from manufacturing date or 500 charge/discharge cycles
Rechargeable lithium-ion battery designed for use with the IntelliVue X3 and IntelliVue MX100 patient monitors. 10.8V/21.6 Wh. Max Charge Current: 2.0A, Max charge Voltage: 12.3V, SMBus 1.1 compliant. 1 each = 1 battery
|Use with philip equipment||867030 (IntelliVue X3), 867033 (IntelliVue MX100)|
|Single- or multi-patient use||Multi-patient|
|Sterile or non-sterile||Non-sterile|
|Use with non-philip equipment||No|
|Package weight||183 gram|
|Battery Life||Minimum 3 years after receipt of product or other factors. See Monitor IFU for further details.|
|WxHxD||69.6 x 72.3 x 21.6 mm (2.7 x 2.8 x 0.8 mm)|
• Discharge 0–60°C (32–140°F)
• Charge 0–60°C (32–140°F)
• Storage and Transport: -20–65°C (-4–149°F)
• Operating: 15–90% Relative Humidity (RH)
• Storage and Transport: 5–95% Relative Humidity (RH)
|Safety||Complies with: UL 62133/IEC 62133|
|Electromagnetic compatibility||Complies with the requirements for FCC Type B computing device, and EN 61000-4-2, and EN 61000-4-3|
|Communication standard||Complies with the SMBus specifi cation v1.1|
When storing the battery, make sure that the battery terminals do not come into contact with conductive substances such as metal objects. For long-term storage of the battery, it is recommended to store it in a cool place at 15 ° C (60 ° C) with a 20% to 40% charge. Storage of the battery in a cool place can slow down the aging process.
Do not store the battery at temperatures other than -20 ° C (-4 ° C) to 60 ° C (140 ° C).
Do not store the battery in direct sunlight. The stored battery should be partially charged to 20-40% capacity every 6 months. Must be fully charged before use.
Note: Prolonged storage of the battery at temperatures above 38 degrees Celsius (100 degrees Celsius) can significantly reduce the expected battery life.
Battery life management
The life of the lithium ion battery depends on the frequency and duration of use. With proper care, it has a lifespan of around 3 years or 500 charge / discharge cycles, whichever comes first. Furthermore, experience has shown that the cumulative stress of daily use increases failure rates as battery life increases. Therefore, it is recommended to replace the Li-ion batteries after 3 years or 500 charge / discharge cycles. The era of lithium-ion batteries began with manufacturing. Check the date of manufacture and the charge and discharge cycles.
1. Select the battery symbol on the patient monitor screen.
2. Select the appropriate battery eject button.
3. If necessary, select the battery eject button again to view the battery details. The screen displays the date of manufacture, the number of charge and discharge cycles and other data about the battery.
If you're interested in 989803196521 philip IntelliVue X3 MX100 Patient monitor Rechargeable Lithium Ion Battery 10.8V 2000mAh ,feel free to contact us .Thank you .
Contact Person: Andy