China World Technology Medical Equipment Service Group
|Brand Name:||Philip Heartstart XL defibrillator|
|Minimum Order Quantity:||1|
|Delivery Time:||3-5 working days|
|Payment Terms:||T/T, Western Union，PayPal|
|Product Name:||Philip Heartstart XL Defibrillator||Type:||M4735A|
|Application:||Medical Equipment,ICU,hospital Device,defibrillator||Warranty:||90days|
|Temperature Operating:||0º C To 55ºC||Temperature Storage:||-20º To 70ºC|
|Humidity:||Up To 95% Relative Humidity||Altitude Operating:||Up To 15,000 Ft.|
|Altitude Storage:||Up To 15,000 Ft.||AC Line Powered::||100-240 VAC, 50/60 Hz, 1.5A (Class 1)|
|Battery Powered:||12 V Rechargeable, SLA||Humidity::||Up To 95% Relative Humidity|
M4735A Defibrillator Machine Parts,
61AY2014 Defibrillator Machine Parts
Philips HeartStart XL M4735A XL+ M3535A M3536A DFM100 defibrillator Energy Select Knob encoder Grayhill 8939 1938 61AY2014 Rev G
This Energy Select Knob is for Philips HeartStart XL M4735A defibrillator.
It’s has Manual mode and AED mode can be enabled at the same time. AED ON position activates AED mode. Manual On Enables manual mode, synchronized cardio version, and pacing (optional).
The HeartStart XL check to see if the pads patient cable and multifunction defibrillator electrode pads are properly connected. If either connection is compromised, you are prompted to fix the problem.
|1||Dark Stripe on Ribbon Cable|
|2||Energy Select Switch|
|3||Square Ribs on Bezel|
|6||Energy Select Knob|
► Removal and Replacement
1. Save the configuration.
If possible, save the customer’s configuration onto a Data Card (or print the configuration) so the configuration can be restored after the repair is complete.
2. Open the case safely.
3. Remove the Energy Select Knob.
Grasp the knob and pull straight out from the Front Bezel.
TIP: To get a better grip on the knob, try wearing rubber gloves.
Disconnect and remove the Energy Select Switch.
a. Unlatch the connector and unplug it from the Keyscan PCA.
b. Using a pliers or wrench, loosen and remove the large nut holding the Energy Select switch into the Bezel. Remove the switch.
TIP: The knob will be difficult to remove. To get a better grip on the knob, try wearing rubber gloves. If needed, use a pair of pliers with soft material over the jaws.
Replace and reconnect the Energy Select Switch.
a. Insert the Energy Select switch into the hole in the Bezel.
b. Fit the switch into place between the square ribs on the Bezel. Be sure the switch is fully bottomed against the Bezel.
c. Replace the lockwasher and nut on the shaft. Tighten the nut securely.
d. Plug the connector into the Keyscan PCA. Be sure it latches.
After this repair is complete, perform the following steps.
1. Replace the Energy Select Knob.
Push the knob onto the shaft. Be sure the knob is pressed fully into place.
2. Reassemble the case.
3. Restore the customer’s configuration.
Enter Configuration Mode and reload the customer’s configuration from the data card, or reconfigure the unit, as needed.
4. Test performance.
In order to avoid causing unnecessary trouble, please read the following carefully.
Please look through all the pictures with patience before placing orders to confirm the product is exactly you want or not as they are also an important part of the product description. What’s more, it is buyer’s responsibility to pay for all the customs fees including all taxes and duties. So it is of great necessity for you to make certain that the equipment you are ordering can be imported in to your country. Once the package leaves our factory, the shipper is not responsible for any delays any more. So please keep it in mind that be sure to check out all the relevant information before placing your orders.
Customer satisfaction. Our biggest pursuit! If you have any questions about Philips HeartStart XL M4735A XL+ M3535A M3536A DFM100 defibrillator Energy Select Knob encoder Grayhill 8939 1938 61AY2014 Rev G, please feel free to contact us at any time, we are glad to answer them.
Contact Person: Rita