China World Technology Medical Equipment Service Group
|Model Number:||M4735A XL|
|Minimum Order Quantity:||1|
|Delivery Time:||3-5 working days|
|Payment Terms:||T/T, Western Union，PayPal|
|Product Name:||Philip M4735A XL Defibrillator||Type:||High Voltage Board|
|Application:||Medical Equipment,ICU,hospital Device||Warranty:||90days|
|Temperature Operating:||0º C To 55ºC||Temperature Storage:||-20º To 70ºC|
|Humidity:||Up To 95% Relative Humidity||Altitude Operating:||Up To 15,000 Ft.|
|Altitude Storage:||Up To 15,000 Ft.||AC Line Powered::||100-240 VAC, 50/60 Hz, 1.5A (Class 1)|
|Battery Powered:||12 V Rechargeable, SLA|
Defibrillator Power PCA Board,
M4735A PCA Board,
M4735-60110 PCA Board
Hospital Medical Equipment Parts Philips HeartStart XL M4735A defibrillator Power PCA Board With Pacer M4735-60110
The Power PCA is located in the bottom case.
Control of defibrillation functions on the Power PCA, including:
a. Initiating a capacitor charge sequence.
b. Monitoring charge on capacitor.
c. Initiating a shock delivery sequence.
Control of pacing functions on the Power PCA, including:
a. Starting/stopping pacing.
b. Controlling Rate and Output as selected by user, and monitoring pacing current delivered.
1. Disconnect the Patient Connector from the Power PCA.
a. Disconnect the 9-pin connector; press on the latch to release.
b. Disconnect the two High Voltage spade connectors by pulling up with a needle nose pliers.
2. Disconnect the Battery PCA from the Power PCA.
a. Disconnect the large white 4-pin connector; press on the latch to release. This connector carries the four battery wires (two red, two black).
b. Disconnect the small 2-pin connector; press on the latch to release.
3. Disconnect the Power Supply.
a. Disconnect the 6-pin connector from the Power PCA; press on the latch to release. Note it only carries three wires.
b. Disconnect the 4-pin connector from the Power PCA; press on the latch to release.
4. Disconnect the Control PCA ribbon cable.
Disconnect the large ribbon cable from the Control PCA. Its other end should already be disconnected from the Control PCA.
5. Remove the Power PCA.
a. Remove the 6-T10 size screws from the Power PCA.
b. Pull the Power PCA up and guide it around the Patient Connector and the case post.
TIP: Lift the back edge of the PCA (nearest the Power Supply) first. Then guide the PCA out to the right to clear the Patient Connector and the case post, and lift the PCA clear of the case.
1. Replace the Power PCA.
a. Ensure the shield is still in its proper position in the bottom case.
b. Guide the Power PCA into position under the Patient Connector.
c. Replace the six T10 screws and tighten.
2. Connect the Control PCA ribbon cable.
Connect the large ribbon cable. Its other end will still be disconnected from the Control PCA.
3. Connect the Power Supply.
a. Connect the 2-pin in-line connector to the Battery PCA; be sure it latches.
b. Connect the 6-pin connector to the Power PCA; be sure it latches.
c. Connect the 4-pin connector to the Power PCA; be sure it latches.
4. Connect the Battery PCA to the Power PCA.
a. Connect the 4-pin connector; be sure it latches.
b. Connect the small 2-pin connector; be sure it latches.
5. Connect the Patient Connector to the Power PCA.
a. Connect the 9-pin connector; be sure it latches.
b. Connect the two High Voltage spade connectors by pushing straight down with a needle nose pliers.
► Noninvasive Pacing
|Waveform:||Monophasic Truncated Exponential|
|Current Pulse Amplitude:||10 mA to 200 mA (5 mA resolution);
accuracy 10 mA to 50 mA ± 5 mA, 50 mA to 200 mA ± 10%
|Pulse Width:||20 ms with accuracy +0, -5 ms.|
|Rate:||30 ppm to 180 ppm (10 ppm increments); accuracy ± 1.5%.|
|Modes:||Demand or Fixed Rate|
|Refractory Period:||340 msec (30 to 80 ppm); 240 msec (90 to 180 ppm).|
Customer satisfaction. Our biggest pursuit! If you have any questions about Philips HeartStart XL M4735A defibrillator Power PCA Board With Pacer M4735-60110, please feel free to contact us at any time, we are glad to answer them.
Contact Person: Rita