China World Technology Medical Equipment Service Group
|Minimum Order Quantity:||1|
|Delivery Time:||3-5 working days|
|Payment Terms:||T/T, Western Union，PayPal|
|Product Superiority:||Original||Key Word:||Battery|
|After-sale Service:||In The Warranty Period And Non-human Damage Can Go To The Sale Free Of Charge For New||Product Status:||In Good Form|
|Company Tenet:||Original Assembled Real Product, Quality Guarantee, Deliver On Time, Preference Price||Stock Status:||Stock Available|
|Minimum Purchasing Quantity:||1unit||AkuPak::||NiCd, 12 V / 1,4 Ah|
|Integrated ::||100 ... 240 Volt, 50 / 60 Hz||Operating Conditions::||0°... 55° C, 30... 95 % Rel. Humidity, But Without Condensation|
|Storage Conditions::||- 20 ... 70 °C, 20 ... 95 % Rel. Humidity,|
240 Volt LiFePO4 Battery,
Primedic LiFePO4 Battery
Primedic HeartSave 6/6S/AED-M/M250/M290/XD10 AkuPak Lite HeartSave LiFePO4 Battery
► Brief Introduction
Rechargeable lithium-ion battery (LiFePO4), rechargeable battery with charging level display and external charging connections. 33-watt power content, nominal operating voltage 13.2 volt continuous current, nominal capacity 2.5 amps.
Rechargeable with DefiMonitor XD product family, charger base and charger clip.
All specifications apply to the new fully charged AkuPak XD device with a temperature of up to 20°C ± 5°C AkuPak LITE/ AkuPak XD: AkuPak LITE/ AkuPak XD usually lasts Up to 4 years or 1000 charging cycles under the following conditions: AkuPak LITE/AkuPak XD is introduced into the device, and the device only remains on standby and is not used, but only performs the usual self-tests on metrax recommended periods. The environmental temperature at all times is 23°C (± 2°C), so that the age is much lower if the devices are stored at external temperatures. Due to the different criteria that may affect the age of AkuPak LITE/AkuPak XD, Metrax assumes no responsibility for the age of AkuPak LITE/AkuPak XD
► Using Multifunction Electrode Pads
To prepare for defibrillation using multifunction electrode pads:
► Preparing for Pacing of HeartStart MRx
If pacing in Demand Mode, apply monitoring electrodes and connect the ECG cable to the HeartStart MRx.
► Charging The AkuPak
The AkuPak can be charged in two ways:
The integrated end-of-discharge detection protects the accumulator from total discharge. The end of charging is detected through the end-of-discharge voltage.
Charging the AkuPak at temperatures other than at the given operating temperatures can damage the accumulator.
A completely empty accumulator must be charged for at least 2 hours. Owing to the type of the accumulator, a very small charging time can lead to errors in interpreting charging by the device. The charging display can show a full accumulator by mistake. Safe operation of the device is also not guaranteed.
Shall the AkuPak not be charged or out of order, the DefiMonitor XD can also be used by means of the integrated mains power supply. The AkuPaks is automatically charged during the use of DefiMonitor XD. The charging time will be prolonged.
In case the device discovers an elevated temperature, it will interrupt the charging process until a normal temperature within a pre-defined value will be measured. This will avoid damaging the AkuPak during charging process.
► Displayed Symbol
|– Self test passed
– Battery / AkuPak capacity >30 %
|– Device is ready for use|
|– Discharged accumulator
– Self test passed
|– Device can be used, the AkuPak might have to be charged or replaced
– Please insert the AkuPak
|– Device might be defective
– Device is defective
|– Please carry out big self test by inserting AkuPak once again or switching on the device once again
– Have the device repaired by your distributor
► About Company:
I have supply the Parts, Accessories and medical equipment of Patient Monitor, Defibrillator Machine, ECG Machine,Ventilator, Anesthesia Machine, Ultrasound such as
Please kindly contact us, if you are interested in Primedic HeartSave 6/6S/AED-M/M250/M290/XD10 AkuPak Lite HeartSave LiFePO4 Battery, Thank you!
Contact Person: Rita