China World Technology Medical Equipment Service Group
|Brand Name:||Nihon Kohden|
|Model Number:||TEC-7621C TEC-7721C|
|Minimum Order Quantity:||1|
|Delivery Time:||3-5 working days|
|Payment Terms:||T/T, Western Union，PayPal|
|Product Name:||Nihon Kohden Cardiolife Defibrillator Main Board Mother Board||Type:||UR-0247 UR-02471(6190-022576K P1)|
|Color:||Green||Dominant Position:||Original,Specialized Engineers With 12 Years Of Experience|
|Condition:||In Good Form||Inventory Situation:||Stock Available|
|Index Terms:||Accessories,Consumable Items||Applications Scenario:||Hospital; Clinic; Hospital Room; ICU;Inpatient Ward|
|Major Material:||Plastic||Function:||In Good Working Condition|
|Aim Of Management:||Quality Guarantee, Deliver On Time, Preference Price||After-sale Service:||In The Warranty Period And Non-human Damage Can Go To The Sale Free Of Charge For New|
TEC-7721C Defibrillator Machine Parts,
TEC-7621C Defibrillator Machine Parts,
Defibrillator Main Mother Board
Nihon Kohden Cardiolife TEC-7621C TEC-7721C defibrillator Main board Mother board PN UR-0247 UR-02471(6190-022576K P1)
When the motherboard is replaced with a new one, make the following adjustments:
- ECG sensitivity and offset
- Charge energy, dellivered energy, TTR
- Battery Charging settings
► Remove the Main board.
After removing the main chassis, you can remove the motherboard.
1. Disconnect the following cables:
To remove the membrane cable from the membrane cable connector, use a small deployment blade driver and fingers to gently release the membrane cable connector lock, as shown in the unfolded image. The card is full because the membrane cable connector is easily damaged.
2. Remove one M3 disc screw, spacer and CNA005 ground lead (connected to AC/DC unit).
3. Remove the three M3 screws from the motherboard.
4. Remove the motherboard.
|Charging Display||Displays the charged energy value on the screen|
|Clock Accuracy||At Surrounding temeprature 25℃(77°F)
At Storage Temperatures -20 to 70 ℃ (-4 to 158°F)
|AC Line Voltage||100 to 240 V|
|AC Line Frequency||50/60 Hz (automatic switching)|
|AC Power Input||Intermittent Load: 450VA or less
Continuous Load: 200VA or less
|DE Power Voltage||12 V|
|DC Power consumptuion||Intermittent Load: 14A or less
Continuous Load: 4.2A or less
|Charging Time||3 Hours or less (with AC, with the power OFF)|
► Delivery Issues
As a general rule, all the orders are going to be shipped within 3 to 5 working days and delivered on time. Of course sometimes there are some special circumstances. For instance, on busy days the shipment will be postponed. But no worries, all the orders will be dispatched finally. We will make great efforts to process the orders and ship them out as soon as possible. Please stay patient and keep in touch with us. If you need the product to be shipped urgently, please make sure you talk to us in advance before placing your orders. In order to make the deal runs successfully, please make it a point and manage your time of placing orders on the basis of your business needs. If you have any questions such as your order is taking much more time than expected or you need to know when will your order arrive, you can contact us at anytime. We will reply you as soon as possible. Your satisfaction is our biggest wish.
Thank you for your time! If you want to explore more details about Nihon Kohden Cardiolife TEC-7621C TEC-7721C defibrillator Main board Mother board PN UR-0247 UR-02471(6190-022576K P1), please let us know. Customer satisfaction. Our biggest pursuit!
Contact Person: Rita