China World Technology Medical Equipment Service Group
|Model Number:||Dash3000 Dash4000 Dash5000|
|Minimum Order Quantity:||1|
|Delivery Time:||3-5 working days|
|Payment Terms:||T/T, Western Union，PayPal|
|Product Name:||GE||Type:||DAS Assembly|
|Application:||Dash3000 Dash4000 Dash5000||Warranty:||90days|
|Functionality:||DAS Assembly||Advantage:||Original,Professional Technical Team|
|Condition:||In Good Working Order||Resolution:||800 By 600 Dpi|
|Machine Type:||Examination Therapy Equipments||TRANSPORTATION:||Express|
|Minimum Order Quantity:||1pcs||Blood Pressure Gain:||10 MV/mmHg ±2%|
|Blood Pressure DC Offset:::||±20 MV (max)||Blood Pressure Noise::||<5 MVp-p (0-300 Hz)|
|Blood Pressure Frequency Response::||Dc To 50 Hz +2/–0 Hz||Blood Pressure Time Delay::||40 Hz Filter, 37 Ms|
Dash5000 Patient Monitor Dash Module,
2006933-002 Patient Monitor Dash Module
GE Dash3000 Dash4000 Dash5000 patient monitor Dash module Masi-mo with DAS Assembly FRU Nibp Pump 2006933-002 E
► Brief Introduction
This DAS Assemble Module is work for GE Dash3000 Dash4000 Dash5000 patient monitor, It’s have Noninvasive Blood Pressure (NBP), Invasive Blood Pressure (BP), ECG, Pulse Oximetry (SPO2), Carbon Dioxide (CO2), Temperature (TEMP).
Perform the AC hi-pot tests for both the ECG and Temp/CO side panel connectors of the device under test. This only needs to be performed when the DAS assembly is opened.
► Replace DAS Assembly
1. Remove the two screws securing the handle to the patient monitor.
2. Remove the monitor assembly from the main unit.
3. Remove the four screws securing the Data Acquisition System (DAS) assembly
4.Remove the DAS assembly by first pulling the left side out 1/4 inch and then sliding the entire assembly out of the patient monitor. If the wireless LAN option is installed, use needle nose pliers to remove the antenna by disconnecting the two coax connectors from the compact flash card.
5. Replace the DAS components.
6. Reassemble the unit in the order of disassembly steps.
7. If you have MasimetSETSPO2 and 4BP Super Stat upgrade installed, please place the tag as shown below.
► NBP pressure range:
|Systolic pressure range||Adult:||30 to 275 mmHg|
|Pediatric:||30 to 235 mmHg|
|Neonatal:||30 to 135 mmHg|
|Diastolic pressure range||Adult:||10 to 220 mmHg|
|Pediatric:||10 to 220 mmHg|
|Neonatal:||10 to 110 mmHg|
|Mean pressure range||Adult:||20 to 260 mmHg|
|Pediatric:||20 to 260 mmHg|
|Neonatal:||20 to 125 mmHg|
► Pulse Oximetry (SPO2)
|Parameters monitored:||Arterial oxygen saturation (SpO2) and periPhilipiliperal pulse rate (PPR)|
|SpO2 range:||50 - 100%|
|PPR range:||30 - 300 beats per minute|
|Actual accuracy depends on probe. Please reference manufacturer’s specifications.|
± 2% (70 - 100% SpO2) ±1 standard deviation
± 3% (50 - 69% SpO2) ±1 standard deviation
± 3 beats per minute
► Packing Details
All machines and units are delivered on pallets or carton so as to make sure the safety to the full extent during the transmission. As a international manufacturer, we deliver medical equipment all over the world, insured and tracked as standard using trusted couriers.
In order to ensure that the goods are delivered to customers in good condition, we will carefully and patiently pack the products. Please rest assured that we will use high-quality packaging materials to pack the goods to avoid being scratched or impacted during transportation. Generally speaking, there is no problem with the goods you receive.
But if you find any problems with the goods during transportation, please contact us as soon as possible. But please pay special attention, if the problem is not caused by transportation, it is another matter. In short, after careful inspection by our engineers and our careful packaging, the goods will eventually arrive in your hands intact.
If you want to explore more details about GE Dash3000 Dash4000 Dash5000 patient monitor Dash module Masi-mo with DAS Assembly FRU Nibp Pump 2006933-002 E , please let us know. Thank you for your time and interest.
Contact Person: Rita