China World Technology Medical Equipment Service Group
|Model Number:||Rainbow RC-12|
|Minimum Order Quantity:||1|
|Delivery Time:||3-5 working days|
|Payment Terms:||T/T, Western Union，PayPal|
|Product Name:||Masi-mo Patient Cable||Type:||Rainbow RC-12|
|Application:||Medical Equipment,ICU,hospital Device||Warranty:||90days|
|Color:||Purple||Power Supply Source:||Electric|
|Rated Voltage:||110V — 240v||Product Conditions:||Running Sweetly|
|Product Characteristics:||Professional, Reliable,Functional,Durable||Post-sales Service:||In The Warranty Period And Non-human Damage Can Go To The Sale Free Of Charge For New|
|Goodness:||New And Original||Supported Languages:||English,French,Italian,Spanish,Japanese,Hindi,Portuguese,Russian,etc|
|Minimum Purchasing Quantity:||1 Pcs||Delivery Terms:||Express|
|Length:||12 Ft /3.7m|
Rainbow Patient Cable,
Mindray Datascope Patient Cable
Masi-mo Rainbow® Patient Cable Rainbow RC-12 for Mindray Datascope DATASCOPE DPM6 DPM7 SPO2 Adapter Cable
Rainbow® RC-12 Patient Cable, 12 ft (3.7m) For use with rainbow and M-LNCS sensors.
It’s 8 pins, purple connector.
This Spo2 Cable is measure SpO2 on a healthy person’s finger and a Pleth wave and PR value are displayed. The displayed SpO2 value is within 95% and 100%.
The SpO2 value displayed on the monitor screen is the average of data collected within a specific time. The shorter the averaging time is, the quicker the patient monitor responds to changes in the patient’s oxygen saturation level. Contrarily, the longer the averaging time is, the slower the patient monitor responds to changes in the patient’s oxygen saturation level, but the measurement accuracy will be improved. For critically ill patients, selecting shorter averaging time will help understanding the patient’s state.
► Setting SpO2 Sensitivity
For Masi-mo SpO2 module, you can set [Sensitivity] to [Normal] or [Maximum] in the [SpO2 Setup] menu. When the [Sensitivity] is set to [Maximum], the patient monitor is more sensitive to minor signals. When monitoring critically ill patients whose pulsations are very weak, it is strongly recommended that the sensitivity is set to [Maximum]. When monitoring neonatal or non-critically ill patients who tend to move a lot, noise or invalid signals may be caused. In this case, it is recommended that the sensitivity is set to [Normal] so that the interference caused by motion can be filtered and therefore the measurement stability can be ensured.
► SpO2 Specifications
|Alarm limit||Range (%)||Step (%)|
|Desat||0 to (high limit – 1)
20 to (high limit – 1)
► Masi-mo SpO2 Module
|Measurement range||1 to 100%|
|Refreshing rate||1 s|
|SpO2 averaging time||2-4 s, 4-6 s, 8 s, 10 s, 12 s, 14 s, 16 s|
|Low perfusion SpO2 accuracy||±2%|
|Low perfusion conditions||Pulse amplitude: >0.02%
Light penetration: >5%
|Accuracy||70 to 100%: ±2% (measured without motion in adult/pediatric mode)
70 to 100%: ±3% (measured without motion in neonate mode)
70 to 100%: ±3% (measured with motion)
0% to 69%: Not specified.
► Measurement Limitations
If you doubt the measured SpO2, check patient vital signs first. Then check the patient monitor and SpO2 sensor. The following factors may influence the accuracy of measurement:
If you have any questions about Masi-mo Rainbow® Patient Cable Rainbow RC-12 , please feel free to contact us at any time, we are glad to answer them. Looking forward to receiving your early inquiry. Thanks for Your Cooperation.
Contact Person: Rita