China World Technology Medical Equipment Service Group
|Minimum Order Quantity:||1|
|Delivery Time:||3-5 working days|
|Payment Terms:||T/T, Western Union，PayPal|
|Product Name:||Philip Toco+ Transducer||Type:||M2735A|
|Application:||Medical Equipment,ICU,hospital Device||Warranty:||90days|
|Material:||Plastic||Product Applications Scenarios:||Outpatient Operating Room,Surgical Operating Room,Emergency Operation Room,etc|
|Preponderance:||Original Assembled Real Product||Sale Protection:||Passion In Serving Our Customer With The Best Service|
|Languages:||Multiple Language Support||Product State:||Work Well|
|Rated Voltage Range:||110V—240v||Power Supply Source:||Electric|
|Size (diameter):||83 Mm/3.27 In||Weight (without Able):||< 220 G/7.8 Oz.|
|Water Ingress Protection Code:||IP 68 (immersion Up To 1 M Water Depth For 5 Hours)|
Philip M2735A Toco Transducer,
Valon F20 Toco Transducer
Philipilip M2735A Toco+ Transducer for Philipilip Avalon F20 FM30 Fetal/Maternal Monitor
► Brief Introduction
Philipilip M2735A Toco+ Transducer can work for Philipilip Avalon F20 FM30 Fetal/Maternal Monitor.
Several parameter frontends are combined on one board. In addition to the Toco frontend, additional supported parameters are DECG, MECG and IUP. A seven-pin ‘D-type’ socket carries all parameter related inputs and outputs. An external mode resistor, connected to one of the pins, automatically detects which mode to set when an adapter cable is plugged in (whether it is DECG, MECG, or IUP).
|Temperature Range||Operating||0°C to 40°C (32°F to 104°F)|
|Storage||-20°C to 60°C (-4°F to 140°F)|
|Humidity Range||Operating||<95% relative humidity @ 40°C/104°F|
|Storage||<90% relative humidity @ 60°C/140°F|
|Altitude Range||Operating||-500 to 3000 m/-1640 to 9840 ft.|
|Storage||-500 to 13100 m/-1640 to 43000 ft.|
► Transducer Cable Assembly
|Transducer Cable (for all fetal transducers)||1|
|Screw Shoulder M2.5||3|
|Screw Cover (pack of 3)||1|
|Transducer Belt Button||1|
► Making Connections
► Testing the Toco+ Transducer (M2735A): DECG Mode
Make a short between the spiral electrode and the reference electrode with your fingers (it is best to wet your fingers first). Use a sterile Fetal Scalp Electrode.
Cut off the plastic tip of the fetal scalp electrode (containing the spiral and reference electrodes) from the end of the wires. Strip the insulation from the end of the wires, and connect them to a patient simulator.
Thank you for your time! If you are interested in Philipilip M2735A Toco+ Transducer for Philipilip Avalon F20 FM30 Fetal/Maternal Monitor, welcome to place your orders. Early order early shipment!
If you have any questions, please feel free to contact us at any time, we are glad to answer them.
Contact Person: Rita