China World Technology Medical Equipment Service Group
|T/T, Western Union,Paypal
|Philip CO2 CapnoTrak Airway Adapter Set
|EsoPhilipageal / Rectal
|3.1 M (10.2')
|Sterile OR Non-Sterile:
EsoPhilipageal Rectal Temperature Probe,
21075A EsoPhilipageal Rectal Temperature Probe,
989803100881 EsoPhilipageal Rectal Temperature Probe
Philipilip 21075A EsoPhilipilipageal Rectal Temperature probe REF 989803100881
Hyperthermia and hypothermia are frequently experienced clinical conditions and their undesirable effects in patients have been well documented in the literature.
Body temperatureis measured continuously using disposable temperature sensors or probes to detect hyper-or hypo-thermia.
These sensors are inserted in the structure designed to fit a specific anatomy of the human body where temperature is measured. The main component of the temperature sensors is a chip which changes resistance with a change intemperature.
The chip used in the EsoPhilipilipageal / Rectal Temperature Probe is a thermistor equivalent to aYSI 400 series. This thermistor is encapsulated in a plasticcap connected to a lead wire. This lead wire has aninsert molded connector on the opposite end to connect with the instrument cable.
The accuracy of the sensor is: ±0.1 ̊C at 37 ̊C and ±0.2 ̊C at 5 ̊C and 45 ̊C.The EsoPhilipilipageal / Rectal Temperature Probe is designed to be inserted into the esoPhilipilipagus, nasoPhilipiliparynx, or rectum. The thermistor assembly is placed inside of a PVC tube size 9 Fr or 12 Fr, with sensing tip placed at the tube’s proximal end. The tube has a bullet tip which allows for atraumatic insertionand is made of medical gradePVC.
The probes are disposable and designed to be used with instruments that are compatible with YSI 400series or equivalent thermistors. The cables to connect the probe to the instrument are Starboard MedicalCable Cat. No. C400-MP-M, C400-MP-MJ, C400-P-M, and C400-P-MJ, or SMITHS Level 1 Cable Cat. No. C400-10. The device is sterile, designed to be disposable, for single useonly. The EsoPhilipilipageal / Rectal temperature probes are supplied in both 9Fr and 12Frsizes.
If the EsoPhilipilipageal / Rectal Temperature probe is utilizedduring surgical proceduresusing electro-cautery, the following may occur:Artificialfluctuations in temperature readings.Localized tissue burns due to the thermistor and lead wire acting as an alternate path for the radio-frequency current to return to ground.The risk of localized tissue burns can be minimized by having the active and ground probes of the electro-cautery system in a close proximity to each other. All parts of the EsoPhilipilipageal / RectalTemperature sensor and cable shallbe away from the electro-cautery probes and connecting cables and, therefore, outside of the radio-frequency current field
Philipilip 21075A EsoPhilipilipageal Rectal Temperature is to be used in routine continuous monitoring of esoPhilipilipageal or rectal temperature as an indicator of the patient's core body temperature.
Suitable for all Philipilip monitors such as models:
MP20, MP30, MP40, MP50
MP60, MP70, MP80, MP90
VM6, VM8, CM10, CM12
Used For Medical machines:
M1029A,M3012A, M3535A/A06, M3536A/A06, M3536M, M8102A, M8105A, M3000A, M3001A, M3002A, M3016A, M3015B, M3015A
|How can I pay for it?
|T/T,Western Union, PayPal are all acceptable.
|What are your main brands?
|Philipilip, Mindray, GE, NIHON KOHDEN, Goldway, etc.
|How long will it take to receive the goods?
|It depends on your shipping address.
|What will you do if you find any quality problems within the warranty period?
|Repair or replace according to the actual situation.
Contact Person: Lucy