China World Technology Medical Equipment Service Group
|Minimum Order Quantity:||1|
|Delivery Time:||3-5 working days|
|Payment Terms:||T/T, Western Union，PayPal|
|Application:||Medical Equipment,ICU,hospital Device||PN:||989803144921|
|Product Situation:||Good In Function||Tag:||Consumable Items,SpO2 Sensor|
|Rated Voltage Forge:||100V—250V||Scenarios:||Hospital, Clinic, Hospital Room, ICU,etc|
|Features Introduction:||Easy To Use,Professional, Reliable,Functional||Dominant Position:||Master The Core Technology|
|Minimum Order Quantity:||One Unit||Sale Protection:||Online Technical Support Services|
|MEANS OF TRANSPORT:||Express Delivery||Length:||1 M (3.3ft)|
|Packaging:||1 Each||Leads:||5 Chest|
Philip M1604A Patient Monitor Accessories,
989803144921 Patient Monitor Accessories
Hospital Medical Patient Monitor Accessories Philipilip M1604A CBL 5 Lead Snap Chest IEC ICU 989803144921
► Brief Introduction
These products are intended for use only with Philipilip devices that are listed as accessories in the device's instruction manual. Additionally, they can be used with the following Philipilip/Agilent/HP devices: 78352C, 78354C, 78834C, M1001A/B, M1002A/B, M3000A, M3002A, M8102A, M8105A, 863063, 863064, 863065, 863066, 863068, 8630.
Electrode placement is color-coded in accordance with IEC 60601-2-27 and AAMI recommendations.
All ECG trunk cables can be used with AAMI or IEC color-coded lead sets. Electrode terminations are available with grappling hooks, snaps and mini clips.
This symbol indicates that the ECG cable, when used in combination with the above-mentioned monitors, is designed to have special protection against electric shock (especially with regard to the permissible leakage current) and is defibrillator-proof.
|5 chest||M1604A||ICU, snap||Reusable||1 m (3.3ft)||1 each||5+5 Trunk cable M1949A, M1663A|
► Electrode-to-Patient Contact
Poor electrode-to-patient contact can result in poor signal quality and relative waveforms, as well as false alarms. Poor contact could result from to poor skin preparation, dried electrode gel, and/or defective ECG cables or lead wires.
Keep in mind the following guidelines to ensure good contact:
► Lead Selection
It is important to select a suitable lead that shows the largest amplitude and cleanest signal so that a QRS complex and R-wave, in particular, can be accurately detected by the monitor. Consider the following lead selection guidelines:
To prevent detection of P-waves or baseline noises as QRS complexes, the minimum detection level for QRS complexes is set at 0.15 mV, according to AAMI-EC 13 specifications.
Customer satisfaction. Our biggest pursuit! If you want to explore more details about Philipilip M1604A CBL 5 Lead Snap Chest IEC ICU 989803144921, please let us know.
Contact Person: Rita