China World Technology Medical Equipment Service Group
|Minimum Order Quantity:||1|
|Delivery Time:||3-5 working days|
|Payment Terms:||T/T, Western Union，PayPal|
|PRODUCT NAME:||Mindray Rechargeable Li-ion Battery||TYPE::||M1943NLLI11S001A|
|APPLICATION::||Mindary DPM2, PM60 Pulse Oximeter||WARRANTY::||90days|
|Key Ward:||Mindray Battery||Product Type:||Battery|
|CONDITION::||In Good Working Order||Rated Voltage:||3.7V|
|Weight:||47g / 1.66oz||Dimensions:||54.00 X 46.25 X 13.00mm|
|Charge Time:||2 Hours To 90% 3.5 Hours To 100%||PN:||M05-010004-08|
Mindray Rechargeable Li Ion Battery,
LI11S001A Rechargeable Li Ion Battery,
M05-010004 Rechargeable Li Ion Battery
Mindray Rechargeable Li-ion battery LI11S001A 3.7V 1800mAh 6.66Wh PN M05-010004-08 for Mindary DPM2, PM60 Pulse Oximeter
► Brief Introduction
Mindray Rechargeable Li-ion battery LI11S001A 3.7V 1800mAh 6.66Wh PN M05-010004-08 is work for Mindary DPM2, PM60 Pulse Oximeter.
If the battery capacity is too low, a technical alarm will be triggered and the [Battery Too Low] message displayed. At this moment, replace the the battery if a lithium ion battery is used. Otherwise, the pulse oximeter will shut down automatically when the battery is depleted. If interrupting patient monitoring is not allowed at this moment, you can place the pulse oximeter in the Charger stand and connect the AC mains. In this case, the pulse oximeter will give prompt message [Battery Type Err] if the alkaline batteries are used. You must pay close attention to the power supply to the pulse oximeter and replace the alkaline batteries or charge the lithium ion battery as soon as interruption of patient monitoring is permissible.
You can charge the lithium ion battery using a Charger stand whether the pulse oximeter is turned on or off. However, monitoring a patient while the battery is being charged is not recommended.
|Weight||47g / 1.66oz|
|Dimensions||54.00 x 46.25 x 13.00mm|
|Run time||24 hours with SpO2 monitored continuously, audio indicators off and backlight brightness set to minimum using a new, fully charged battery at ambient temperature 25℃.|
|Charge time||2 hours to 90%
3.5 hours to 100%
|Shutdown delay||Max. 10 minutes after the low battery alarm first occurs.|
► When a lithium ion battery is used, the battery icon indicates the battery status as follows:
Indicates that the battery works correctly. The solid portion represents the current charge level of the battery in proportion to its maximum charge level.
Indicates that the battery has low charge level and need to be charged.
Indicates that the battery is almost depleted and need to be charged immediately.
► Conditioning the Lithium Ion Battery
A lithium ion battery needs at least two conditioning cycles when it is put into use for the first time. A battery conditioning cycle is one complete, uninterrupted charge of the battery, followed by a complete, uninterrupted discharge of the battery. A lithium ion battery should be conditioned regularly to maintain its useful life. Condition a battery once when it is used or stored for two months, or when its run time becomes noticeably shorter.
To condition a lithium ion battery, follow this procedure:
Now, the battery is conditioned and the pulse oximeter can be returned to service.
If you have other questions about Mindray Rechargeable Li-ion battery LI11S001A 3.7V 1800mAh 6.66Wh PN M05-010004-08 for Mindary DPM2, PM60 Pulse Oximeter, please feel free to contact us. We are always ready to serve you.
Contact Person: Rita