China World Technology Medical Equipment Service Group
|Place of Origin:||China|
|Minimum Order Quantity:||1pcs|
|Delivery Time:||3-5 working days|
|Payment Terms:||T/T(wire transfer), Western Union, Paypal|
|Product Name:||13-100-0013 Biolight BLT Q5 Patient Monitor Mainboard||Brand:||Biolight|
|Part No .:||13-100-0013||Display Size:||12.1"|
|Resolution:||800 X 600||Size:||318mm X 264mm X 152mm|
|Weight:||4.5kg||Power Voltage:||AC 100-240Volt|
|Battery:||Rechargeable Lithium Battery||Operating Time:||>210 Minutes Under Normal Conditions|
|Alarm:||Level – Low, Medium, High||Alarm Indications:||Auditory And Visual|
|NIBP Measurement Reviewing:||1000 Groups||Full Disclosure Waveform:||96 For 3 Waveforms (with 4G SD Card)|
Q5 Patient Monitor Mainboard,
13-100-0013 Patient Monitor Mainboard
13-100-0013 Biolight BLT Q5 patient monitor mainboard Maintenance
Note: To ensure the functionality and safety of the device, it should be checked after one year of use.
Seasonal Security Checks:
The power plug should be cleaned at least once a year. Too much dust on the plug can cause a fire. The following safety inspections must be performed by qualified personnel with sufficient training, knowledge, and practical experience to perform these tests at least every 12 months. The data must be written to the device registry. If a device malfunctions or fails any of the following tests, it must be repaired.
① Check that there is no mechanical or functional damage to the device and accessories.
② Check whether the security label can be read.
③ Check whether the fuse matches the rated current and breaking characteristics.
④ Check if the device works normally as described in the user manual.
⑤ Check the protective earth resistance according to IEC 60601-1: 1988 + A1: 1991+ A2: 1995: Limit value: 0.1W.
⑥ Leakage current test according to IEC 60601-1: 1988 + A1: 1991 + A2: 1995: Limits: NC 500 μA, SFC: 1000 μA.
⑦ Test case leakage current according to IEC 60601-1: 1988 + A1: 1991 + A2: 1995: Limits: NC 100 μA, SFC: 500 μA.
⑧ IEC patient leakage current test (normal operation)60601-1:1988 + A1:1991 + A2:1995: Limit: type BF: for a.c.: 100μA, for d.c.: 10μA. type CF: for a.c.: 10μA, for d.c.: 10μA.
⑨ Test the patient leakage current under single fault condition according IEC 60601-1:1988 + A1:1991 + A2:1995: Limit: type BF: for a.c.: 500μA, for d.c.: 50μA. type CF: for a.c.: 50μA, for d.c.: 50μA.
⑩ Test the patient leakage current Mains voltage on applied part: According IEC 60601-1:1988 + A1:1991 + A2:1995: Limit: type BF: for a.c.: 5mA. type CF: for a.c.: 50uA
Biolight BLT Q5 patient monitor mainboard 13-100-0013
|Use place||Hospital Devices; Medical Care; ICU; Medical Equipments and Accessories|
|Deliver time||3-5 days|
Protect your device and accessories from dust and dirt.
damaged equipment Follow these rules:
Dilute according to manufacturer's recommendations or use the lowest possible concentration.
2. Do not immerse any part of the device in liquid.
3. Do not spill liquids on the device or accessories.
4. Do not allow liquids to enter the house.
Do not use abrasive cleaners (like steel wool or silver polish) or abrasive cleaners (like acetone or acetone).
Contact Person: Andy