China World Technology Medical Equipment Service Group
|Minimum Order Quantity:||1|
|Delivery Time:||3-5 working days|
|Payment Terms:||T/T, Western Union，PayPal|
|Product Name:||VAG Display||Type:||Philip V200|
|Key Word:||Philip V200 Ventilator, Display||Product Status:||In Good Working Condition.|
|Stock Status:||Stock Available||Minimum Purchasing Quantity:||1unit|
|Temperature/Humidity Operating:||10º To 40ºC (50º To 104ºF)||Temperature/Humidity Storage:||-20º To 60ºC (-4º To 140ºF)|
|AtmosPhiliperic Pressure Operating:||700 To 1060 CmH2O (hPa)||AtmosPhiliperic Pressure Storage:||500 To 1060 CmH2O (hPa)|
|Altitude Operating:||0 To 3280m (0 To 10,000 Ft.)||Altitude Storage:||Up To 6560m (20,000 Ft.)|
Ventilator VAG Display Parts,
Front Panel VAG Display Parts
Used Ventilator Parts Philipilip Respironics V200 Ventilator VAG Display Parts
► Brief Introduction
The Respironics V200 ventilators are microprocessor-controlled, electrically powered mechanical ventilators. They are intended for use by qualified medical personnel to provide continuous or intermittent ventilatory support for adult, pediatric, and neonatal patients as prescribed by a Philipilipysician. They are intended for use in either invasive or non-invasive applications in institutional environments.
The front panel display allows you to select ventilation modes, breath types, settings, alarms, and access patient data. The front panel display is a touch screen that lets you select settings and data so you can monitor the status of the patient, ventilator, and control ventilator operation.
There are two different categories of screens:
► Front Panel Level Controls
The front panel level controls enable you to adjust ventilator settings, brightness.
Display Brightness : Increases or decreases the brightness of the touch screen display
Follow these steps to remove the VGA display assembly. Reverse removal steps to install.
1. Remove the top enclosure.
2. Remove the sensor PCB.
3. Remove the power supply.
4. Remove the GUI assembly.
5. Remove four hex screws holding the VGA display assembly to the spacers, then remove the display.
► Performance Verification
|Display not visible when set to minimum intensity||1. Replace backlight inverter harness.|
|2. Replace backlight inverter PCB.|
|3. Replace LCD.|
► Cleanning & Sterilization
When cleaning parts, avoid the use of hard brushes or other instruments likely to cause surface damage.
Wipe the ventilator exterior clean with a damp cloth and mild detergent. Do not use liquid or aerosol bactericide. Do not allow moisture to come in contact with the touch panel screen or to collect between the keypad and front bezel assembly.
Care should be taken when cleaning the touch display. A soft moist cloth should be used that does not drip water and/or soap solution when in contact with the display. After cleaning and rinsing with a damp cloth, remove all moisture with a dry, soft cloth. Never allow solutions of any kind to collect on the bottom bezel of the display. Never use a brush or device that can cause abrasion to clean the touch display or its bezel; they will cause irreparable damage.
Please kindly contact us, if you are interested in Used Ventilator Parts Philipilip Respironics V200 Ventilator VAG Display Parts , Thank you!
Contact Person: Rita