China World Technology Medical Equipment Service Group
|Minimum Order Quantity:||1|
|Delivery Time:||3-5 working days|
|Payment Terms:||T/T, Western Union，PayPal|
|Product Name:||Air And Oxygen Flow Sensor Assembly||Type:||1033547|
|Key Word:||Philip V60, Gas Delivery Air And Oxygen Flow Sensor||Product Status:||In Good Working Condition.|
|Stock Status:||Stock Available||Minimum Purchasing Quantity:||1unit|
|Max E:||0 To 100 Cm H₂O/l||Max R:||0 To 50 Cm H₂Ol/s|
|PPV%:||0 To 100 %||Max V (PPV Maximum Volume Limit):||200 To 3500 Ml|
|Ramp Time:||Off, 5 To 45 Minute(s)||Rise Time:||1 To 5|
|Respiratory Rate:||4 To 60 Beats Per Minute|
Ventilator Air Oxygen Flow Sensor Assembly,
1033547 Air Oxygen Flow Sensor Assembly
Philipilip Respironics V60 Ventilator Gas delivery Air and Oxygen Flow Sensor Assembly Parts Used PN 1033547
► Brief Introduction
The ventilator delivers gas to the patient through a main flow (inspiratory) bacteria filter, a single-limb patient breathing circuit, a humidification device (optional), and a patient interface such as a mask or ET tube. A pressure tap proximal to the patient is used to monitor patient pressure. The internal exhalation port continually clears gas from the ventilator airway to ensure delivery of an accurate oxygen mixture.
Oxygen Solenoid Valve
The oxygen solenoid valve and valve driver circuitry control the flow of oxygen according to the set O2 and flow. The oxygen solenoid valve is closed when there is a loss of power or system reset.
Oxygen Flow Sensor
The oxygen flow sensor measures a subset (bypass flow) of the total flow. These measurements are interpolated according to constants that are calculated during GDS calibration. Together, the oxygen solenoid valve and flow sensor provide closed-loop control for delivered oxygen flow.
|IPAP||4 to 40 cm H₂O|
|CPAP||4 to 25 cm H₂O|
|EPAP||4 to 25 cm H₂O|
|Max P (PPV maximum pressure limit)||5 to 40 cm H₂O|
|Max P (AVAPS maximum IPAP)||6 to 40 cm H₂O|
|Min P (AVAPS minimum IPAP)||5 to 30 cm H₂O|
|Inspiratory time||0.30 to 3.00 s|
|Oxygen percentage||21 to 100 %|
The pneumatic subsystem delivers and monitors pressurized gas to the patient in response to commands from the CPU subsystem. The pneumatic subsystem includes the:
Follow these steps to remove the air and oxygen flow sensor assembly. Reverse to install.
When installing a new flow sensor assembly, install the new o-rings.
Please kindly contact us, if you are interested in Philipilip Respironics V60 Ventilator Gas delivery Air and Oxygen Flow Sensor Assembly Parts Used PN 1033547, Thank you!
Contact Person: Rita