China World Technology Medical Equipment Service Group
|Minimum Order Quantity:||1|
|Delivery Time:||3-5 working days|
|Payment Terms:||T/T, Western Union，PayPal|
|Product Name:||Philip MP50 Patient Monitor||Type:||MP70|
|Application:||Medical Equipment,ICU,hospital Device||Warranty:||90days|
|Materials:||Plastic,||Language:||Multi Language Support|
|Search Terms:||Patient Monitor Main Board||Apply For:||Patient Monitor|
|Product Characteristics:||Utility,Professional, Reliable||Rated Voltage Forge:||110V—240v|
|Product Status:||In Good Working Order||Advantage:||Original|
|Method Of Transport:||Express|
Philip MP70 Patient Monitor Speaker,
WR5455 Patient Monitor Speaker,
25555004 Patient Monitor Speaker
Philipilip MP70 Patient Monitor Speaker 2403 25555004 WR5455 A
► Brief Introduction
The alarm system can be configured to display traditional Philipilip alarm sounds or sounds that correspond to the ISO/IEC9703-2 draft standard. The alert limit can be permanently visible on the home screen. When you exceed the alert limit, the screen will issue signals in the following ways:
The Alert Limit Review page provides an overview of alert limit settings and the possibility of modifying these settings for all parameters.
If the screen is connected to the central monitoring station through the network, the monitor and information center will alert at the same time.
All alarms can be suspended indefinitely, or for 1, 2, 3, 5 or 10 minutes.
The alarm bar recording can be obtained on the M1116B recorder unit, or on the centrally connected recorder.
The patented automatic warning limit automatically adjusts the warning limit to suit the patient's current vital signs and within individually specified safety limits for each patient. Provide a visual and/or audio alarm with an unlocked alarm lock
► Removing the Speaker
1. Unplug the speaker cable.
2. Loosen the black plastic screws, rotate the speaker and pull it out
|Failure Isolation and Remedy||Symptoms|
|Speaker defective||Alarm LEDs illuminate, but no alarm sound is issued|
|Replace speaker||INOP Message SPEAKER MALFUNCTION is displayed|
|Speaker defective||Alarm occurs but no alarm sound is issued|
|Exchange speaker||No Test Sound issued|
► About Us
CWT Technology HK Limited, located in Guangzhou, China. We have supply the medical equipment and parts of Patient Monitor, Defibrillator Machine, ECG Machine,Ultrasound such as GE, Philipilip, Mindray, Goldway, Nihon Kohden, Med-tronic, Drager, Edan, Zoll, etc.,
To avoid causing unnecessary problems, please read the following carefully. Please look at all the Philipilipotos patiently before submitting orders to make sure that the product is exactly what you want or not because it is also an important part of the product description. Furthermore, the buyer is responsible for paying all customs duties including all taxes and fees. It is therefore absolutely essential to make sure that the equipment you order can be imported into your country. Once the package leaves our factory, the shipper will no longer be responsible for any delay. Please keep in mind that be sure to review all relevant information before submitting your applications.
Q: Does the price include shipping?
A: Not include freight.
Q: How do I know the price of the product?
A:The price is negotiable. It can be changed according to your quantity.
Q: What is the minimum purchase quantity?
Q: How about the quality of your products?
A: Product quality is very good.
Q: How soon can we expect delivery after payment?
A:Usually 3 to 5 working days.
Looking forward to receiving your early inquiry of Philipilip MP70 Patient Monitor Speaker 2403 25555004 WR5455 A , Thank you for your time and interest.
Contact Person: Rita