China World Technology Medical Equipment Service Group
|Minimum Order Quantity:||1pcs|
|Delivery Time:||3-5 working days|
|Payment Terms:||T/T(wire transfer), Western Union, Paypal|
|Product Name:||Mindray Imec12 Patient Monitor Recoder Printer||Brand Name:||Mindray|
|MOQ:||One Piece||Warranty:||90 Days|
|Delivery Method:||DHL,UPS ,Fedex ,etc||Delivery Time:||3-5 Working Days|
|Application:||Hospital Devices; Medical Care; ICU; Medical Equipments And Accessories||Payment Method:||T/T ,Paypal ,Western Union|
|Payment Terms:||In Full Before The Shipment|
Imec12 Patient Monitor Recorder Printer,
Mindray Patient Monitor Recorder Printer
Mindray Imec12 patient monitor recoder printer with good condition for Hospital
Mindray Imec12 patient monitor recoder printer for Hospital
The recorder receives the data from the main board, and then sends it to the thermal print head for printing. There is a hard key (start/stop recording) and a green LED on the front of the recorder. It is connected to the motherboard
|Power interface||Introduces a DC from the main board|
Convert the input power to a voltage suitable for each module, and then forward them to each module.
The figure below shows how it works
|CPU||Control the communications between modules|
|Signal interface||Control the communications between the main board and the recorderCPU|
|Motor drive circuit||Receives the control signals from the CPU and then forwards them to thestep engines|
|Key&LED interface||Includes a button and an LED which are directly controlled by the CPU|
Recorder Verification Tool Required: None.
1. Print the ECG waveform. The recorder should print correctly and the printout should be clear.
2. Configure the recorder for out of paper and other issues and the monitor should be providing the appropriate information. The burner should function normally after the problem is resolved.
3.Open the automatic alarm record of each parameter, and then set the limit of each parameter outside the set alarm limit. When a parameter alarm occurs, the corresponding alarm recording should be triggered
To enable/disable the recorder, select [Recorder] and toggle between [On] and [Off].
The recorder is disabled if [Recorder] is switched off in the menu
1.Push the two clips to one side and take out the drive board of the burner, as shown in the figure below
2.Unscrew the two PT2X6 screws, and disconnect the cable and flexible cord between the recorder drive board and the recorder keyboard, as shown in the figure below. Then remove the thermal print head and recorder drive board.
3.Unscrew the two PT2X6 screws shown in the figure below and take out the keyboard
|Symptoms||Possible Cause||Corrective Action|
1.Check if the recorder status LED lights
2. If yes, enable the module in menu. Otherwise, check for other possible causes
|Paper reversed||Re-install the paper roll.|
|Cable defective orpoorly connected||
1.Check that cable between recorder and main board isproperly connected
2. Check that connecting cables and connectors are notdamaged
|Recorder powersupply failure||Check if the power module outputs 5 V DC and 12V DCcorrectly|
|Recorder failure||Replace the recorder.|
|Poor printquality orPrint head dirtypaper notfeedingproperly||Paper roll notproperly installed||Stop the recorder and re-install the paper roll|
|Print head dirty||
1.Check the thermal print head and the paper roller forforeign matter.
2. Clean the thermal print head with an appropriate cleansolution
|Print head failure||Replace the print head.|
|Recorder failure||Replace recorder|
Contact Person: Andy