China World Technology Medical Equipment Service Group
|Brand Name:||Mindray D3 Rechargeable Li-ion battery|
|Minimum Order Quantity:||1|
|Delivery Time:||3-5 working days|
|Payment Terms:||T/T, Western Union，PayPal|
|Product Name:||Mindray D3 Rechargeable Li-ion Battery||Type:||M3516A 989803107041|
|Charge Time:||Less Than 3 Hours To 90% And Less Than 4 Hours To 100% With Equipment Power Of||Warranty:||90days|
|Capacity Indicator:||5-segment Led Indicator For Fast Battery Capacity Evaluation||Capacity:||Monitoring Mode: 6 Hours, Monitoring With A 5-lead ECG,Resp, SpO , CO And NIBP Measurements Set At An Interval Of 15 Minutes|
|Wi-Fi:||Disabled||Energy Accuracy::||±2 J Or 15% Of Setting, Whichever Is Greater, Into 50 Ohm|
|Power On Time::||Less Than 2 Seconds With A New, Fully Charged Battery||Charge Time::||Ess Than 7 Seconds To 360 Joules With A New, Fully Charged Battery|
|ECG Recovery Time::||Less Than 2.5 Seconds||Patient Impedance Range::||25 To 300 Ω (External Defibrillation)|
|Output Energy::||1,2,3,4,5,6,7,8,9,10,15,20,30,50,70,100,150,170,200,300,360 J||Current::||1.8 To 0.8 A|
|Frequency::||50/60 Hz (±3 Hz)||Input Voltage::||12 VDC|
|Power Consumption::||190 W|
D3 Defibrillator Rechargeable Li Ion Battery,
LI24I001A Defibrillator Rechargeable Li Ion Battery,
Li Ion Battery Defibrillator Machine Parts
Hospital Medical Equipment Parts Mindray D3 defibrillator Rechargeable Li-ion battery Model No.LI24I001A
► Brief Introduction
The equipment is designed to operate on battery power when external power supply is not available. The battery is charged whenever the equipment is connected to AC mains or the DC power supply through an external DC/AC adapter, regardless of whether or not the equipment is currently turned on. In case of power failure, the equipment will automatically run power from internal batteries. So we recommend you always install a fully charged battery in the equipment.
The equipment is configured with one smart lithium ion batteries which are free of maintenance.
On-screen battery symbols indicate the current battery charge status, for an example:
≤100%, but >80% of capacity
≤80%, but >60% of capacity
≤60%, but >40% of capacity
≤40%, but >20% of capacity
≤20% of capacity
Low battery and charging is required immediately
Battery is not installed
You can also check the battery's charge status by pressing the fuel gauge button on the battery to illuminate the battery gauge. The fuel gauge consisting of 5 LEDs, each LED represents a charge of approximately 20% of capacity. If the battery charge is too low, a technical alarm will be triggered and the "Low Battery" message displayed in the Technical Alarm Area. At this moment, change the battery or apply external power to the equipment.
Always connect the equipment to AC mains whenever it is possible.
Always install a fully charged battery in the equipment.
After long term use, the power capacity indicated by the battery symbol may be different from the actual capacity.
Always observe the alarm information displayed on the screen
► Battery Run time
|Monitoring||≥2.5 h||Without recording, typical ECG monitoring, LCD brightness set to 1|
|Defibrillation||≥100 discharges||360J discharges at intervals of one minute, without recording|
|pacing||≥2 h||50 Ω load impedance, pacing rate: 80bpm,pacing output 60mA, without recording|
|Battery type||14.8V/3Ah, smart lithium ion battery, rechargeable and free of maintenance, one battery can be configured|
|Charge time||Less than 2 hours to 80% and less than 3 hours to 100% with equipment power off;
Less than 3.5 hours to 80% and less than 4.5 hours to 100% with equipment power on.
|Battery fuel gauge||5 LEDs indicating the current battery charge level|
|Shutdown delay||At least 20 minutes of monitoring and six 360J discharges (after the low battery alarm occurs)|
► Charging batteries
The batteries can be charged only when they are installed in the equipment or using an device approved by the equipment manufacturer. With the equipment turned off and at a temperature of 25°C (77°F), a completely discharged battery charges to 80% of its capacity in approximately 2 hours, and to 100% of its capacity in approximately 3 hours. Batteries charge at a lower rate with the equipment turned on.
Batteries should be charged at temperatures between 0°C (32°F) to 45°C (113°F). To optimize performance, a fully (or nearly fully) discharged battery should be charged as soon as possible
► Low Battery Alarm
If the equipment is run on battery and the battery charge is low, a technical alarm "Low Battery" will be triggered. In this case, replace the battery with a fully charged on or connect the equipment to external power supply immediately.
If the battery is almost depleted, a prompt "Battery depleted! System will shut down imminently. Connect to AC mains or replace battery" pops up. In monitoring, manual defibrillation, and pacing mode, additional alarm lights and alarm tones are provided. In this case, take appropriate actions immediately. This prompt will not disappear until the battery is replaced or the equipment is connected to the external power supply. The equipment automatically shuts down if no action is taken within a period of about 3 minutes.
The Low Battery alarm means that the battery is beginning to weaken and should be replaced at the first opportunity. At least 20 minutes of monitoring and six full energy shocks can be performed when the Low Battery alarm is activated. Replace the battery or connect the equipment to AC mains as soon as possible.
Customer satisfaction. Our biggest pursuit! If you have any questions about Mindray D3 defibrillator Rechargeable Li-ion battery Model No.LI24I001A, please feel free to contact us at any time, we are glad to answer them.
Contact Person: Rita