China World Technology Medical Equipment Service Group
|Brand Name:||Philip Heartstart M4735A XL Battery|
|Model Number:||M3516A 989803107041|
|Minimum Order Quantity:||1|
|Delivery Time:||3-5 working days|
|Payment Terms:||T/T, Western Union，PayPal|
|Product Name:||Philip Heartstart M4735A XL Battery||Type:||M3516A 989803107041|
|Application:||Medical Equipment,ICU,hospital Device,defibrillator||Warranty:||90days|
|Temperature Operating:||0º C To 55ºC||Temperature Storage:||-20º To 70ºC|
|Humidity:||Up To 95% Relative Humidity||Altitude Operating:||Up To 15,000 Ft.|
|Altitude Storage:||Up To 15,000 Ft.||AC Line Powered::||100-240 VAC, 50/60 Hz, 1.5A (Class 1)|
|Battery Powered:||12 V Rechargeable, SLA|
M4735A XL Defibrillator SLA Battery,
M3516A Defibrillator SLA Battery,
989803107041 Defibrillator SLA Battery
Hospital Medical Equipment Parts Philipilip M4735A XL Defibrillator High Voltage Module Board
The M4735A is powered by a rechargeable Sealed Lead Acid (SLA) battery (M3516A). Proper care of these batteries will ensure that they have the energy required to operate the M4735A and deliver the appropriate therapy. The M3516A battery differs from similar batteries in that it is capable of providing much higher current without causing an internal protection device to open and disconnect the battery.
The defibrillator will take longer to charge when powered by AC without a battery installed. To ensure optimal performance, always have a fully charged battery in the defibrillator, even when using AC power.
The repair PhilipilipilosoPhilipilipy for the SLA battery (M3516A) is unit replacement. If a battery fails, it is replaced, not repaired.
► Power Indicator Matrix
|AC POWER||BATTERY CHARGING||Connected to AC Mains?||Battery in Place?||Unit Power ON?|
|G = GREEN
OR = ORANGE/AMBER
|Y = YES N = NO --- = don’t care|
|G||OR||Y||Y||---||Battery is charging.|
|G||G||Y||Y||---||Battery at least 90% charged.|
►Battery Capacity Test
The HeartStart XL uses the M3516A Battery Pack. It is rechargeable sealed lead acid battery. Battery maintenance begins when you receive a new battery and continues throughout the life of the battery.
As sealed lead acid batteries age due to usage and storage conditions, their capacity to store and deliver energy diminishes. The Battery Capacity Test provides a quantitative measure of battery capacity. Performing a Battery Capacity Test periodically over the life of the battery gives you an indication of how that battery is aging, and what level of performance it is capable of delivering.
A battery’s capacity is most affected by the duration of use and how deeply it is discharged before recharging, Therefore, some general guidelines for performing the Battery Capacity Test are as follow:
► To perform a Battery Capacity Test(CT).
Battery Capacity Test Results
|Elapsed Time≥ 95 minutes
Low Battery time ≥ 10 minutes
|1. The battery passed the test
2. Record "Pass CT" and the date on the bottom of the battery
3. Recharge the battery before use.
|Elapsed Time＜ 95 minutes
Low Battery time ＜ 10 minutes
|1. The battery failed the test.
2. Record "Fail CT" and the date on the bottom of the battery.
3. Discard the battery appropriately.
Customer satisfaction. Our biggest pursuit! If you have any questions about Philipilip Heartstart M4735A XL defibrillator M3516A Sealed Lead Acid SLA Battery REF 989803107041, please feel free to contact us at any time, we are glad to answer them.
Contact Person: Rita