China World Technology Medical Equipment Service Group
|Model Number:||M4735A XL|
|Minimum Order Quantity:||1|
|Delivery Time:||3-5 working days|
|Payment Terms:||T/T, Western Union，PayPal|
|Product Name:||Philip M4735A XL Defibrillator||Type:||High Voltage Board|
|Application:||Medical Equipment,ICU,hospital Device||Warranty:||90days|
|Temperature Operating:||0º C To 55ºC||Temperature Storage:||-20º To 70ºC|
|Humidity:||Up To 95% Relative Humidity||Altitude Operating:||Up To 15,000 Ft.|
|Altitude Storage:||Up To 15,000 Ft.||AC Line Powered::||100-240 VAC, 50/60 Hz, 1.5A (Class 1)|
|Battery Powered:||12 V Rechargeable, SLA|
M4735A XL Defibrillator Machine Parts,
Module Board Defibrillator Machine Parts,
M4735A XL High Voltage Board
Hospital Medical Equipment Parts Philipilip M4735A XL Defibrillator High Voltage Module Board
The M4735A HeartStart XL Defibrillator/Monitor is for used in the hospital by qualified medical personnel trained in the operation of the device and qualified by training in basic life support, advanced cardiac life support, or defibrillation. It mush be used by or on the order of Philipilipysician.
When operating as semi-automatic external defibrillator in AED Mode, the M4735A HeartStart XL Defibrillator/Monitor is suitable for use by medical personnel trained in basic life support that includes the use of an AED.
When operating as a defibrillator/monitor in Manual Mode, the M4735A HeartStart XL Defibrillator/Monitor is suitable foe use by healthcare professionals trained in advanced cardiac life support.
►Delivered Energy Accuracy:
|Selected Energy||Delivered Energy (J)||Accuracy|
|Load Impedance (Ohms)|
1. Replace the high voltage board
a. Ensure the shield is still in its proper position in the bottom case.
b. Guide the high voltage board into position under the Patient Connector.
c. Replace the six T10 screws and tighten.
2. Connect the high voltage board ribbon cable.
Connect the large ribbon cable. Its other end will still be is connected from the high voltage board.
3. Connect the Power Supply.
a. Connect the 2-pin in-line connector to the Battery PCA; be sure it latches.
b. Connect the 6-pin connector to the high voltage board; be sure it latches.
c. Connect the 4-pin connector to the high voltage board; be sure it latches.
4. Connect the Battery PCA to the high voltage board.
a. Connect the 4-pin connector; be sure it latches.
b. Connect the small 2-pin connector; be sure it latches.
5. Connect the Patient Connector to the high voltage board.
a. Connect the 9-pin connector; be sure it latches.
b. Connect the two High Voltage spade connectors by pushing straight down with a needle nose pliers.
The two spade connectors are different sizes to ensure correct connection.
► After Repair
After this repair is complete, perform the following steps.
1. Connect and replace the defibrillator capacitor.
2. Reassemble the case.
3. Restore the customer’s configuration.
Enter Configuration Mode and reload the customer’s configuration from the data card, or reconfigure the unit as needed.
4. Test performance.
Conduct Performance Verification Testing as described in Performance
►Packing and Delivery Clauses
All machines and devices are transported in pallets or cartons to ensure maximum safety during transportation. As an international manufacturer, we provide medical equipment all over the world and use reliable messengers as a standard for insurance and tracking. For us, it is very important to deliver your products to you accurately, quickly and safely.
We have been working hard to provide the best shipping service. The shipping cost largely depends on your location, what you ordered and how fast you want your package to arrive. If you have any questions, please feel free to contact us.
As for delivery time, we will do our best to process the order and ship it as soon as possible. Please be patient and stay in touch with us. Generally speaking, most orders will be shipped within 3-5 working days and can arrive on time. However, on busy days, the shipment time may be delayed by 5 working days. If you need your items urgently, please be sure to confirm with us before placing your order.
Where is the place of shipment?
Guangdong Province, China.
What about your engineers?
Our engineers are very professional and have 12 years of working experience.
What is your mode of transportation?
|Q4||How do I know the price of the product?|
The price is negotiable. It can be changed according to your quantity.
Customer satisfaction. Our biggest pursuit! If you have any questions about Philipilip M4735A XL Defibrillator High Voltage Module Board, please feel free to contact us at any time, we are glad to answer them.
Contact Person: Rita