China World Technology Medical Equipment Service Group
|Place of Origin:||China|
|Minimum Order Quantity:||1pcs|
|Delivery Time:||3-5 working days|
|Payment Terms:||T/T(wire transfer), Western Union, Paypal|
|Product Name:||Zoll 3lead ECG Patient Cable (12Ft)PN 8000-0026 New Original||Brand:||Zoll|
|REF:||8000-0026||Condition:||New & Original|
|Number Of Channels:||3||Country/Region Of Manufacture:||China|
|MOQ:||One Piece||Warranty:||90 Days|
PN 8000-0026 ECG Patient Cable,
12Ft ECG Patient Cable,
20517621019 ECG Patient Cable
Zoll 3lead ECG patient cable (12Ft)PN 8000-0026 new original
Quick Judgement and Maintenance of Lead Wire Failure of ECG
During the use of the electrocardiograPhilipilip, poor contact or disconnection of a certain lead may affect the normal tracing of the electrocardiogram due to factors such as the lead wire is often pulled and twisted. It is a common type of electrocardiograPhilipilip failure. Now summarizes the method of using the electrocardiograPhilipilip itself to judge the fault.
Trouble Philipilipenomenon: abnormal or serious interference of the electrocardiogram waveform of a certain lead or all leads occurs when doing an electrocardiogram for a normal person.
Failure analysis: The lead wires of the electrocardiograPhilipilip are composed of four red, yellow, blue, and black limbs and five chest leads. When it works, it sends the electrocardiograPhilipilip signal obtained on the electrode to the input end of the electrocardiograPhilipilip. For tracing after magnification. According to the principle of ECG generation and the connection method of the electrodes, use Table 1 to find out a certain faulty lead wire.
Troubleshooting: After finding the faulty lead wire, first determine its fault location. Most of the faults are near the lead wire electrode, and the Philipilipenomenon is that when the hand is pulled there, there is a sense of extension. When touched by the hand, it is softer and thinner than other places. For this reason, it can be above the fault point. After disconnecting at ~5cm, unplug the core wire and electrode welding or core wire and core wire, shield and shield welding, pay attention to the shield not to connect with the core wire. Then use insulating tape or put on the heat-shrinkable heat-shrinkable insulation welding joint before welding. Turn it on again, and all the leads of the ECG are normal.
|Item name||Zoll 3lead ECG patient cable (12Ft)PN 8000-0026 new original-|
|Use place||Hospital Devices; Medical Care; ICU; Medical Equipments and Accessories|
|Deliver time||3-5 days|
Model number for each brands are as follows :
Patient monitor :B105 ,B125,B450,B650 ,B20i ,B30 ,DASH3000,DASH4000,DASH5000,DASH2500
ECG/EKG :MAC800 ,MAC1200,MAC400 ,MAC5000 ,MAC5500
Patient monitor :MP20,MP30,MP40,MP50,MP60,MP70,MP80,MP90,VM4,VM6,VM8, MX600, MX700, MX800 ,MX400,MX450
Fetal monitor :FM20,FM30
Defibrillator :M3543A,M3535A,M3536A.Heartstart XL+
Patient monitor :MEC-1000,MEC-2000,IMEC10,IMEC12,BeneVision N1
BeneView T5 ,BeneView T6,BeneView T8
ECG/EKG:ECG 1250P,ECG-1250A,ECG-1350,cardiofax S,cardiofax Q
Defibrillator:M series ,E series
Defibrillator :Lifepack 12 ,lifepack 20 ,lifepack 1000
Masi-mo SET Radical-7 pulse oximeter,Masi-mo 9 pin Adult Finger clip Sensor Model LNCS DCO PN 1863,Masi-mo rainbow Rc-12 patient cable
For models ,spare parts and accessories without listing above ,feel free to contact us .We have sufficient stock for used medical device.
Contact Person: Andy