China World Technology Medical Equipment Service Group
|Minimum Order Quantity:||1|
|Packaging Details:||By Carton|
|Delivery Time:||3-5 Working Days|
|Payment Terms:||T/T, Western Union, Paypal|
|Product Name::||Thermal Array Recorder||MODEL NO.::||M1116B|
|MOQ::||1 Pc||WARRANTY::||90 Days|
|APPLICATION::||Philip MP50 Patient Monitor||COLOR::||White|
|MATERIAL::||PCB,Plastic,etc.||SHIPPING TIME::||3-5 Days|
|TRANSPORTATION::||Air Express||PAYMENT::||T/T, Western Union, Paypal|
|CONDITION::||Good Working Condition||Weight:||507.5 G / 17.9 Oz|
|Size:||73 X 99.6 X 97 Mm / 2.9 X 3.6 X 3.9 In.|
Philip MP50 Patient Monitor Parts,
M1116B Patient Monitor Parts,
M1116B Thermal Array Recorder
Philipilip MP50 Patient Monitor Thermal Array Recorder M1116B Printer Recorder
► Brief Introduction
The M1116B recorder module offers thermal recording capability for the Philipilip patient monitoring family, including the V24/V26 and the Philipilip Intellivue system. M1116B recorder, it can write up to three superimposed waveforms and three annotation lines. The Philipilip M1116B module offers ten speeds.
The recording service retrieves current and stored Philipilipysiological and state data to format a continuous tape recording. A recording can be triggered manually by the operator or automatically by an alarm condition. The registration service uses the services of the recording interface to control an M1116B recorder. The registration service can also send data to a central recorder.
► The printer interface manager provides a high-level interface to a printer. It provides the means to:
1. Establish a connection to the printer
2. Transfer the data to the printer
3. Get printer status
4. Close the connection to the printer
The printer interface manager also monitors the connection to the printer and whether the printer accepts data (eg, more paper). The printer interface manager informs the operator in such cases.
► Cleaning recorder printer
Long periods of recording at low speeds (1 or 2cm/min) can accumulate paper waste on the print head and cause uneven thinness of the horizontal stripes.
1. Remove the recorder.
2. Open the door of the recorder and remove the paper screw from behind the rubber roller.
3. Tear off the excess paper or roll it up into the roll chamber so it doesn't get in the way.
4. Pass a cloth cleaning strip instead of paper through the rubber roller until about 2 inch strips come out of the top of the roller.
5. Close the door of the recorder and align the ends of the strip with the top of the door.
6. Pull the strip out of the recorder, holding the top edge of the cleaning strip between the thumb and the index finger.
7. Open the door and make sure the paper cavity is dust-free. Please throw away the paper and replace the recorder.
|Temperature Range||Operating||+5 to 45 °C (41 to 113°F)|
|Non-operating||-10 to 70 °C (14 to 158 °F)|
|Humidity Range||Operating||95% Relative Humidity (RH) max @ 40°C (104°F) (non condensing)|
|Non-operating||95% Relative Humidity (RH) max @ 65°C (150°F) (non condensing|
|Altitude Range||Operating||up to 3048 m (10000 ft)|
|Non-operating||up to 3048 m (10000 ft)|
► Delivery Issues
As a rule, all orders are sent within 3-5 business days and delivered on time. If you need urgent delivery, please feel free to contact us in advance before placing orders. If you have any questions, for example, if your order takes much longer than expected or you need to know when it will be delivered, you can contact us at any time. We will reply as soon as possible. Your satisfaction is our greatest desire.
► Contact us
In order to successfully complete the transaction, please contact us in time if you have any questions. Company or product if you're telling details or unclear about the warranty period of shrine or if our products and service anything about all good Suggestions call, if you have any good suggestions for our products and services, you can contact us to raise them.
If you are interested in Philipilip MP50 Patient Monitor Thermal Array Recorder M1116B Printer Recorder, Please feel free to contact us.
Contact Person: Rita